Feasibility and reference values of left atrial longitudinal strain imaging by two-dimensional speckle tracking
- Matteo Cameli†1,
- Maria Caputo†1,
- Sergio Mondillo1,
- Piercarlo Ballo2Email author,
- Elisabetta Palmerini1,
- Matteo Lisi1,
- Enzo Marino3 and
- Maurizio Galderisi4
© Cameli et al; licensee BioMed Central Ltd. 2009
Received: 22 December 2008
Accepted: 08 February 2009
Published: 08 February 2009
The role of speckle tracking in the assessment of left atrial (LA) deformation dynamics is not established. We sought to determine the feasibility and reference ranges of LA longitudinal strain indices measured by speckle tracking in a population of normal subjects.
In 60 healthy individuals, peak atrial longitudinal strain (PALS) and time to peak longitudinal strain (TPLS) were measured using a 12-segment model for the left atrium. Values were obtained by averaging all segments (global PALS and TPLS) and by separately averaging segments measured in the two apical views (4- and 2-chamber average PALS and TPLS).
Adequate tracking quality was achieved in 97% of segments analyzed. Inter and intra-observer variability coefficients of measurements ranged between 2.9% and 5.4%. Global PALS was 42.2 ± 6.1% (5–95° percentile range 32.2–53.2%), and global TPLS was 368 ± 30 ms (5–95° percentile range 323–430 ms). The 2-chamber average PALS was slightly higher than the 4-chamber average PALS (44.3 ± 6.0% vs 40.1 ± 7.9%, p < 0.0001), whereas no differences in TPLS were found (p = 0.93).
Speckle tracking is a feasible technique for the assessment of longitudinal myocardial LA deformation. Reference ranges of strain indices were reported.
The left atrium plays as a booster pomp during late ventricular diastole, as a reservoir for the inflow volume received from pulmonary veins during ventricular systole and isovolumic relaxation, and as a passive conduit during early ventricular diastole and diastasis [1, 2]. Although estimates of left atrial (LA) function can be obtained by two-dimensional echocardiography, Doppler analysis of transmitral and pulmonary vein flow, and Tissue Doppler (TD) assessment of LA myocardial velocities [3–5], quantification of effective LA function still remains a challenging issue. Assessment of atrial deformation profiles obtained using Doppler-derived strain imaging has been recently proposed as an alternative method of exploring LA function. However, this approach is limited by a number of potential drawbacks, including suboptimal reproducibility, angle dependence, and the confounding effect of noise artifacts.
Two-dimensional strain imaging is an echocardiographic technique that uses standard B-mode images for speckle tracking analysis. The speckle pattern (acoustic backscatter generated by the reflected ultrasound beam) is followed frame-by-frame, using a statistical approach based on the detection of the best matching area. The displacement of this speckled pattern is considered to follow myocardial movement, and a change between speckles is assumed to represent myocardial deformation[8, 9].
The aims of this study were to define the feasibility of speckle tracking-based strain imaging for the evaluation of LA wall deformation in a population of healthy subjects, and to identify normality ranges for corresponding strain values.
Sixty consecutive adult healthy subjects, referring to our Echo Laboratory for a diagnostic examination, were included in the study group. All had unremarkable clinical history and normal findings at physical examination, ECG, and baseline echocardiography, and none of them was taking cardiac medications. All subjects gave their written informed consent for participation in the study.
Echocardiographic studies were performed using a high-quality echocardiograph (Vivid 7, GE, USA). Subjects were studied in the left lateral recumbent position. Measurements of left ventricular (LV) and LA dimensions were made in accordance with current American Society of Echocardiography recommendations. LV ejection fraction was measured using the modified biplane Simpson's rule. The ratio between peak early (E) and late (A) diastolic LV filling velocities and E wave deceleration time were determined by standard Doppler imaging. The timings of mitral and aortic valve opening and closure were defined by pulsed wave Doppler tracings of mitral inflow and LV outflow.
For speckle tracking analysis, apical four- and two-chamber views images were obtained using conventional two-dimensional gray scale echocardiography, during breath hold with a stable ECG recording. Particular attention was given to obtain an adequate gray scale image, allowing reliable delineation of myocardial tissue and extracardiac structures. Three consecutive heart cycles were recorded and averaged. The frame rate was set between 60 and 80 frames per second. These settings are recommended to combine temporal resolution with adequate spatial definition, and to enhance the feasibility of the frame-to-frame tracking technique.
Reproducibility of PALS and TPLS measurements was assessed in 20 randomly selected subjects. Intra and inter-observer variability coefficients were calculated using images independently recorded in two different occasions by the same investigator or by two different observers.
Data are shown as mean ± SD. Inter- and intra-observer reproducibility was assessed by calculating variability coefficients. Reference values were expressed as mean ± SD and 5–95° percentile ranges. Comparisons were performed using the Student's t test for paired data. A P value < 0.05 was considered statistically significant. Analyses were performed using the SPSS (Statistical Package for the Social Sciences, Chicago, Illinois) software Release 11.5.
General characteristics of the study population
32.8 ± 13.6
Female gender (n)
171.3 ± 8.1
66.7 ± 9.8
Body surface area (m2)
1.73 ± 0.4
Body mass index (kg/m2)
22.6 ± 2.2
Systolic blood pressure (mmHg)
120 ± 10.7
Diastolic blood pressure (mmHg)
77.3 ± 5.2
Heart rate (bpm)
73.3 ± 10.0
End-diastolic LV diameter (mm)
44.9 ± 4.8
End-systolic LV diameter (mm)
28.3 ± 4.2
LV ejection fraction (%)
60.0 ± 4.3
Left atrial diameter (mm)
28.6 ± 4.8
Left atrial area (cm2)
13.5 ± 2.5
Mitral E/A ratio
1.54 ± 0.4
Deceleration time (ms)
283.3 ± 61.0
When the reproducibility was separately considered in the two apical views, inter-observer variability coefficients were 4.3% and 4.6% for 4- and 2-chamber average PALS, and 5.4% and 5.3% for TPLS, respectively. Corresponding intra-observer variability coefficients were 3.6%, 4.0%, 4.5%, and 4.8%.
Reference ranges of left atrial strain indices
Mean ± SD
5–95° Percentile range
42.2 ± 6.1
40.1 ± 7.9
44.3 ± 6.0
368.0 ± 29.9
364.2 ± 42.6
367.4 ± 34.1
In this study, speckle tracking imaging was found to be a feasible and reproducible method to assess LA longitudinal strain in healthy subjects. The reproducibility of measurements was good, with lower variability in comparison with that obtained by Doppler-derived LA strain imaging. These data suggest that speckle tracking may be considered a promising tool to explore LA myocardial deformation dynamics. Data observed in our population were also used to provide reference ranges for speckle-based LA strain indices.
In our population, average peak longitudinal strain was higher in apical 2-chamber view than in apical 4-chamber view. It can be hypothesized that the confounding effect of atrial septum and pulmonary veins ostia in the 4-chamber view – two zones where strain profile is abnormal – may have played a role in determining this discrepancy. In contrast, the average time to peak longitudinal strain was similar in the two apical views. Interestingly, these findings are in contrast with a recent report on LA strain measured by the Doppler-derived technique, which found similar peak strain values among different LA walls, but different timing measures. These discrepancies should be evaluated in further studies. It should be noted that in contrast to Doppler-derived strain imaging, speckle tracking has the advantage of being angle-independent, and to be less affected by reverberations, side lobes and drop out artifacts. Nonetheless, intrinsic limitations of speckle tracking include strict frame rate dependency, potential errors in epicardial/endocardial border tracing in subjects with suboptimal image quality, and need for an appropriate learning curve to achieve adequate experience in using analysis softwares.
Some limitations should be considered. The potential difficulty of accurately obtaining a region of interest close enough to the effective shape of the left atrium, and the risk of contamination by signal components arising from structures surrounding the left atrium should also be considered. Although the post-processing time in this study was relatively short, it closely depends on the sonographer's experience. Lastly, because a dedicated software for LA strain analysis has not yet been released, we used the current software for LV analysis to study the LA pattern strain. Future evolutions in this regard may be useful to improve tracking quality of LA myocardial deformation, and to provide a better instrument for the study of LA function. Lastly, it should be emphasized that as for other echocardiographic new technologies, speckle tracking is progressively entering the clinical practice despite no definite data regarding reference ranges and no clear demonstration of clear additive value in particular clinical conditions. The results of this study may contribute to partially fill this gap giving insight on the potential application of speckle tracking to the study of LA function, but further larger analyses are needed.
In summary, speckle tracking can be considered a feasible and reproducible technique for the assessment of LA longitudinal deformation dynamics. Normal values of longitudinal strain indices were reported.
The authors wish to thank Vittorio Rastelli and Fabio Cattaneo, GE Healthcare, Milano, Italy, for their important technical support.
- Pagel PS, Kehl F, Gare M, Hettrick DA, Kersten JR, Warltier DC: Mechanical function of the left atrium. New insights based on analysis of pressure-volume relations and Doppler echocardiography. Anestesiology. 2003, 98: 975-994. 10.1097/00000542-200304000-00027.View ArticleGoogle Scholar
- Di Salvo G, Galderisi M, Rea A, Ansalone G, Dini FL, Gallina S, Mele D, Montisci R, Sciomer S, Mondillo S, Di Bello V, Marino PN, Gruppo di Lavoro di Ecocardiografia; Società Italiana di Cardiologia: Evaluation of atrial function by echocardiography. G Ital Cardiol. 2007, 8: 225-235.Google Scholar
- Nakao F, Wasaki Y, Kimura M, Iwami T, Iida H, Wakeyama T, Miura T, Ogawa H, Matsuzaki M: Evaluation of left atrial function by the functional volume change curve derived from Doppler flow spectra. Jpn Circ J. 2001, 65: 953-7. 10.1253/jcj.65.953.View ArticlePubMedGoogle Scholar
- Marino P, Faggian G, Bertolini P, Mazzucco A, Little W: Early mitral deceleration and left atrial stiffness. Am J Physiol Heart Circ Physiol. 2004, 287: H1172-1178. 10.1152/ajpheart.00051.2004.View ArticlePubMedGoogle Scholar
- Pérez-Paredes M, Gonzálvez M, Ruiz Ros JA, Giménez DM, Carnero A, Carrillo A, Cubero T, Martínez-Corbalán FR, García Almagro F: Assessment of Left Atrial Wall Velocities by Pulsed Wave Tissue Doppler Imaging. A New Approach to the Study of Atrial Function. Rev Esp Cardiol. 2004, 57: 1059-1066. 10.1157/13068167.View ArticlePubMedGoogle Scholar
- Di Salvo G, Caso P, Lo Piccolo R, Fusco A, Martiniello AR, Russo MG, D'Onofrio A, Severino S, Calabró P, Pacileo G, Mininni N, Calabró R: Atrial myocardial deformation properties predict maintenance of sinus rhythm after external cardioversion of recent-onset lone atrial fibrillation: a color Doppler myocardial imaging and transthoracic and transesophageal echocardiographic study. Circulation. 2005, 112: 387-395. 10.1161/CIRCULATIONAHA.104.463125.View ArticlePubMedGoogle Scholar
- D'hooge J, Heimdal A, Jamal F, Kukulski T, Bijnens B, Rademakers F, Hatle L, Suetens P, Sutherland GR: Regional strain and SR measurements by cardiac ultrasound: principles, implementation and limitations. Eur J Echocardiogr. 2000, 1: 154-170. 10.1053/euje.2000.0031.View ArticlePubMedGoogle Scholar
- Teske AJ, De Boeck BW, Melman PG, Sieswerda GT, Doevendans PA, Cramer MJ: Echocardiographic quantification of myocardial function using tissue deformation imaging, a guide to image acquisition and analysis using tissue Doppler and speckle tracking. Cardiovascular Ultrasound. 2007, 5: 27-10.1186/1476-7120-5-27.View ArticlePubMedPubMed CentralGoogle Scholar
- Notomi Y, Lysyansky P, Setser RM, Shiota T, Popović ZB, Martin-Miklovic MG, Weaver JA, Oryszak SJ, Greenberg NL, White RD, Thomas JD: Measurement of Ventricular Torsion by Two-Dimensional Ultrasound Speckle Tracking Imaging. J Am Coll Cardiol. 2005, 45: 2034-2041. 10.1016/j.jacc.2005.02.082.View ArticlePubMedGoogle Scholar
- Di Salvo G, Drago M, Pacileo G, Rea A, Carrozza M, Santoro G, Bigazzi MC, Caso P, Russo MG, Carminati M, Calabro' R: Atrial function after surgical and percutaneous closure of atrial septal defect: a strain rate imaging study. J Am Soc Echocardiogr. 2005, 18: 930-933. 10.1016/j.echo.2005.01.029.View ArticlePubMedGoogle Scholar
- D'Andrea A, Caso P, Romano S, Scarafile R, Cuomo S, Salerno G, Riegler L, Limongelli G, Di Salvo G, Romano M, Liccardo B, Iengo R, Ascione L, Del Viscovo L, Calabrò P, Calabrò R: Association between left atrial myocardial function and exercise capacity in patients with either idiopathic or ischemic dilated cardiomyopathy: A two-dimensional speckle strain study. Int J Cardiol.
- Lang RM, Bierig M, Devereux RB, Flachskampf FA, Foster E, Pellikka PA, Picard MH, Roman MJ, Seward J, Shanewise JS, Solomon SD, Spencer KT, Sutton MS, Stewart WJ, Chamber Quantification Writing Group; American Society of Echocardiography's Guidelines and Standards Committee; European Association of Echocardiography: Recommendations for Chamber Quantification: A Report from the American Society of Echocardiography's Guidelines and Standards Committee and the Chamber Quantification Writing Group, Developed in Conjunction with the European Association of Echocardiography, a Branch of the European Society of Cardiology. J Am Soc Echocardiogr. 2005, 18: 1440-1463. 10.1016/j.echo.2005.10.005.View ArticlePubMedGoogle Scholar
- Serri K, Reant P, Lafitte M, Berhouet M, Le Bouffos V, Roudaut R, Lafitte S: Global and regional myocardial function quantification by two dimensional strain. J Am Coll Cardiol. 2006, 47: 1175-1181. 10.1016/j.jacc.2005.10.061.View ArticlePubMedGoogle Scholar
- Langeland S, D'hooge J, Wouters PF, Leather HA, Claus P, Bijnens B, Sutherland GR: Experimental validation of a new ultrasound method for the simultaneous assessment of radial and longitudinal myocardial deformation independent of insonation angle. Circulation. 2005, 112: 2157-2162. 10.1161/CIRCULATIONAHA.105.554006.View ArticlePubMedGoogle Scholar
- Kaluzynski K, Chen X, Emelianov SY, Skovoroda AR, O'Donnell M: Strain rate imaging using two-dimensional speckle tracking. IEEE Trans Ultrason Ferroelectr Freq Control. 2001, 48: 1111-1123. 10.1109/58.935730.View ArticlePubMedGoogle Scholar
- Sirbu C, Herbots L, D'hooge J, Claus P, Marciniak A, Langeland T, Bijnens B, Rademakers FE, Sutherland GR: Feasibility of strain and strain rate imaging for the assessment of regional left atrial deformation: a study in normal subjects. Eur J Echocardiogr. 2006, 7: 199-208. 10.1016/j.euje.2005.06.001.View ArticlePubMedGoogle Scholar
- Leung DY, Boyd A, Ng AA, Chi C, Thomas L: Echocardiographic evaluation of left atrial size and function: current understanding, pathophysiologic correlates, and prognostic implications. Am Heart J. 2008, 156: 1056-64. 10.1016/j.ahj.2008.07.021.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.