This article has Open Peer Review reports available.
Hand-carried ultrasound performed at bedside in cardiology inpatient setting – a comparative study with comprehensive echocardiography
© Tsutsui et al; licensee BioMed Central Ltd. 2004
Received: 26 October 2004
Accepted: 17 November 2004
Published: 17 November 2004
Hand-carried ultrasound (HCU) devices have been demonstrated to improve the diagnosis of cardiac diseases over physical examination, and have the potential to broaden the versatility in ultrasound application. The role of these devices in the assessment of hospitalized patients is not completely established. In this study we sought to perform a direct comparison between bedside evaluation using HCU and comprehensive echocardiography (CE), in cardiology inpatient setting.
We studied 44 consecutive patients (mean age 54 ± 18 years, 25 men) who underwent bedside echocardiography using HCU and CE. HCU was performed by a cardiologist with level-2 training in the performance and interpretation of echocardiography, using two-dimensional imaging, color Doppler, and simple calliper measurements. CE was performed by an experienced echocardiographer (level-3 training) and considered as the gold standard.
There were no significant differences in cardiac chamber dimensions and left ventricular ejection fraction determined by the two techniques. The agreement between HCU and CE for the detection of segmental wall motion abnormalities was 83% (Kappa = 0.58). There was good agreement for detecting significant mitral valve regurgitation (Kappa = 0.85), aortic regurgitation (kappa = 0.89), and tricuspid regurgitation (Kappa = 0.74). A complete evaluation of patients with stenotic and prosthetic dysfunctional valves, as well as pulmonary hypertension, was not possible using HCU due to its technical limitations in determining hemodynamic parameters.
Bedside evaluation using HCU is helpful for assessing cardiac chamber dimensions, left ventricular global and segmental function, and significant valvular regurgitation. However, it has limitations regarding hemodynamic assessment, an important issue in the cardiology inpatient setting.
KeywordsHand-carried ultrasound comprehensive echocardiography inpatient setting
Bedside echocardiography can bring important anatomical and hemodynamic information for the management of critically ill patients, and is often required in hospitalized patients for the assessment of left ventricular function. Standard echocardiographic equipments, while optimal, have large size and sometimes are difficult to maneuver in the emergency room or intensive care units. Recently, hand-carried ultrasound (HCU) devices have been demonstrated to broaden the versatility in ultrasound application. Due to their portability and low cost, HCU acts like a stethoscope, providing information beyond physical examination at the point-of-care [1, 2]. Although it has been shown to improve the detection of cardiovascular abnormalities over the physical examination, its role in the assessment of hospitalized patients is not completely established [3, 4]. This study was undertaken to compare the findings of the bedside evaluation using HCU to the comprehensive echocardiography (CE), in cardiology inpatient setting.
54 ± 18
Acute coronary syndrome
Postoperative of cardiac surgery
Total atrioventricular conduction block
All patients underwent two echocardiographic evaluations. First, HCU was performed with the portable device OptiGo (Philips Medical Systems, Andover, Massachusetts, USA) and, consecutively, by a commercially available system (HDI 5000, Philips Medical Systems, Bothell, Washington, USA) equipped with a 4-2 MHz transducer and second-harmonic imaging. HCU was performed by one same cardiologist with level 2-training in the performance and interpretation of echocardiography according to the specifications of the American Society of Echocardiography , after a period of instructions about the HCU settings. CE was performed by one experienced echocardiographer with level-3 training , and was considered the gold standard. Each investigator was blinded to the results of the other examination. The final echocardiographic report was based on the results of CE.
OptiGo is equipped with a 2.5 MHz phased-array transducer and operates on a rechargeable Lithium ion battery, which facilitates its use at bedside. HCU was performed using two-dimensional imaging, color Doppler flow mapping, and simple caliper measurements. Images were frozen and scrolled for review and the measurements were performed on-line.
CE evaluation included two-dimensional with second-harmonic imaging, M-mode, and both spectral and color Doppler flow mapping. Images were recorded on videotape or digitalized. The aorta, left atrium, left ventricular end-diastolic and end-systolic diameters, as well as interventricular septal and posterior wall thickness were measured according to the recommendations of the American Society of Echocardiography . The left ventricular ejection fraction (LVEF) was visually estimated, and a normal ventricular function was defined as LVEF ≥ 55%. According to the segmental wall motion analysis, patients were divided as those with segmental wall motion abnormality (WMA) and those without WMA. Pericardial effusions were classified as mild, moderate, or large effusion.
Valve structure and function were analyzed. Dysfunctions were classified into mild, moderate, or severe degree according to the qualitative evaluation by HCU, and using both qualitative and quantitative parameters by CE . A significant valvular regurgitation was defined in our study as those of moderate or severe degree . On the other hand, non-significant valvular regurgitation was defined as those of no, trace, or mild degree. The estimation of transvalvar gradients and valvular areas in patients with prosthetic and stenotic valves, as well as the estimation of pulmonary artery pressure, were performed only by CE .
The intraobserver variability of CE findings was assessed in 15 randomly assigned patients, with analysis made at least 4 weeks apart. The interobserver agreement between the experienced echocardiographer and the level 2-trained cardiologist was assessed by the analysis of 15 CE recorded examinations.
Continuous data are expressed as mean ± one standard deviation (SD) and categorical data as proportions. Comparisons between groups for continuous variables were made using Student t test. Chi-square and Fisher Exact tests were used for categorical variables. Agreement between HCU and CE results were assessed by the Kappa statistics. Interobserver and interobserver variability was determined by intra-class correlation and linear regression. A two-tailed p value <0.05 was considered significant.
Determination of cardiac chamber dimensions and left ventricular function
Cardiac chamber measurements obtained by comprehensive echocardiography (CE) and by hand-carried ultrasound (HCU)
28.7 ± 4.1
27.8 ± 4.2
Left atrium (mm)
45.5 ± 7.7
44.0 ± 7.5
57.1 ± 11.2
54.9 ± 10.7
44.4 ± 15.5
42.8 ± 10.7
10.0 ± 2.4
9.4 ± 2.4
9.6 ± 1.8
8.7 ± 1.7*
47 ± 16
44 ± 15
The analysis of segmental wall motion was not possible by HCU in two patients, due to poor endocardial border delineation, and was deemed feasible in all patients using CE. In the remaining 42 patients, HCU correctly identified eight of the 11 patients with WMA by CE, and failed to identify three of them. These three false-negative results occurred in patients with bad image quality by HCU. On the other hand, among the 31 patients without WMA by CE, HCU correctly identified 27 patients, and had four false-positive results. Among these four cases, three patients had global left ventricular dysfunction and one had asynchronic movement of the interventricular septum. The agreement between HCU and CE for the detection of WMA was 83% (Kappa = 0.58; p < 0.001).
CE identified nine patients with pericardial effusion, one patient with a moderate effusion and the remaining eight patients with mild effusions. HCU correctly detected and classified pericardial effusion in seven patients, and failed to diagnose two patients with mild effusions, both localized posterior to the left ventricle.
Analysis of valvular dysfunction
Significant mitral valve regurgitation was diagnosed by CE in 16 patients, and by HCU in 13 of them. The agreement between HCU and CE for detection of this abnormality was 93% (Kappa = 0.85; p < 0.001). Significant aortic valve regurgitation was detected in three patients by CE and in two by HCU, and the agreement between the two techniques was 98% (Kappa = 0.89; p < 0.001). Significant tricuspid regurgitation was detected by CE in 16 patients and by HCU in 12 of them. However, there was one additional case misdiagnosed by HCU as of significant degree. The agreement between HCU and CE for the detection of significant tricuspid regurgitation was 88% (Kappa = 0.74; p < 0.001).
Four patients (9%) had aortic valve stenosis, two cases of moderate, one of mild, and the other one of severe degree. Qualitative analysis of valve structure by HCU was capable to identify aortic stenosis in two of these patients, but the severity of these lesions was not determined. There were six (14%) patients with prosthetic valves and eight (18%) patients with pulmonary hypertension, in which estimation of transvalvar gradients or valvular areas were performed only by CE. A complete evaluation of these patients was not possible using HCU due to its technical limitations in determining hemodynamic parameters.
Intra and interobserver variability
The intraobserver agreement for detection of pericardial effusion, WMA, and significant valvular dysfunction was 100%, and the interobserver agreement was 91%. There was an excellent correlation between LVEF estimated in the first and second evaluation by the same observer (r = 0.91; p < 0.05). The correlation between LVEF estimated by the experienced echocardiographer and the cardiologist with level 2-training in echocardiography was r = 0.88 (p < 0.05).
The present study describes the value of bedside evaluation of cardiac patients using HCU in comparison to CE. Our results demonstrate that HCU may be used for the assessment of cardiac chamber dimensions, estimation of left ventricular function, and detection of WMA. Moreover, we found a good agreement between HCU and CE for detecting significant valvular regurgitation. However, HCU presents important limitations regarding the assessment of prosthetic and stenotic valves, as well as for the evaluation of patients with pulmonary hypertension, due to the lack of spectral Doppler.
Bedside echocardiography is a frequently used diagnostic tool in the cardiology inpatient setting and can affect the patient management, direct further diagnostic work-up, and modify therapeutic decisions. The recent development of portable ultrasound devices has the potential to allow quick and easy-to-use echocardiography at the point-of-care, although its value in hospitalized patients was not completely defined. In patients with cardiovascular disorders, HCU has been shown to increase the diagnostic accuracy over physical examination when performed by cardiologists with level-2 training in echocardiography . In the present study we confirmed the usefulness of these portable devices for estimating cardiac chamber dimensions and left ventricular function, which are frequent indications for bedside echocardiographic examination. We also demonstrated a good agreement between CE and HCU for detecting WMA. We would like to emphasize that CE with second-harmonic imaging was chosen as the gold standard for evaluation of segmental wall motion, since it has already been proven that this imaging modality ameliorates the endocardial border delineation [9, 10]. Another point to be noted is that our population included a high proportion of patients with left ventricular global dysfunction, in whom detection of segmental abnormality can be somehow challenging by non-experienced observers. In at least three of the four false-positive results, the presence of global myocardial impairment due to cardiomyopathy could lead to a confounding diagnosis of segmental impairment. In the false-negative cases, the use of second-harmonic modality improved the quality of imaging, allowing for a better visualization of endocardial thickening and detection of segmental left ventricular dysfunction. Our results are in accordance with recently published data demonstrating that HCU was highly concordant with clinical diagnosis of acute coronary syndrome based on the analysis of wall motion by these portable devices .
However, when considering the full spectrum of abnormalities in hospitalized patients, previous reports demonstrated that hand-carried bedside echocardiography failed to quantify valvular regurgitation and also missed findings relevant to clinical questions in a significant number of patients. These studies concluded that HCU falls far short of standard echocardiography in evaluation of critically ill patients [3, 12]. We do believe that the lack of spectral Doppler in HCU is an important limitation for evaluation of cardiac patients, since this technique provides unique hemodynamic information, especially in patients with prosthetic or stenotic valves, and pulmonary hypertension. In our study population, these clinical conditions occurred in a considerable number of cases, since we included patients in the pre and postoperative period of valvular surgery.
In the present study, we did not evaluate the effect of HCU on patient management. Agreement between CE and HCU for detection of WMA was analyzed in 42 patients, since two patients were initially excluded because of inadequate acoustic window. The specific issue of agreement between HCU and CE according to image quality was not addressed in the present study because of the limited number of patients in each group.
Conclusions and clinical implications
We concluded that HCU is useful for bedside assessment of left ventricular global and segmental left ventricular function as well as for evaluation of significant valvular regurgitation. However, it has limitations regarding hemodynamic assessment, which is an important issue in the cardiology inpatient setting.
Therefore, we emphasize that bedside echocardiography using HCU should be performed by cardiologist with at least level 2-training in echocardiography, and always complemented with CE when hemodynamic evaluation is necessary.
- Spencer KT, Anderson AS, Bhargava A, Bales AC, Sorrentino M, Furlong K, Lang RM: Physician-performed point-of-care echocardiography using a laptop platform compared with physical examination in the cardiovascular patient. J Am Coll Cardiol 2001, 37: 2013-2018. 10.1016/S0735-1097(01)01288-8View ArticlePubMedGoogle Scholar
- Vourvouri EC, Poldermans D, Schinkel AF, Koroleva LY, Sozzi FB, Parharidis GE, Bax JJ, Roelandt JR: Left ventricular hypertrophy screening using a hand-held ultrasound device. Eur Heart J 2002, 23: 1516-1521. 10.1053/euhj.2002.3162View ArticlePubMedGoogle Scholar
- Goodkin GM, Spevack DM, Tunick PA, Kronzon I: How useful is hand-carried bedside echocardiography in critically ill patients? J Am Coll Cardiol 2001, 37: 2019-2022. 10.1016/S0735-1097(01)01293-1View ArticlePubMedGoogle Scholar
- Kimura BJ, Amundson SA, Willis CL, Gilpin EA, DeMaria AN: Usefulness of a hand-held ultrasound device for bedside examination of left ventricular function. Am J Cardiol 2002, 90: 1038-1039. 10.1016/S0002-9149(02)02699-1View ArticlePubMedGoogle Scholar
- Quinones MA, Douglas PS, Foster E, Gorcsan JIII, Lewis JF, Pearlman AS, Rychik J, Salcedo EE, Seward JB, Stevenson JG, Thys DM, Weitz HH, Zoghbi WA: ACC/AHA clinical competence statement on echocardiography: a report of the American College of Cardiology/American Heart Association/American College of Physicians-American Society of Internal Medicine Task Force on Clinical Competence. J Am Coll Cardiol 2003, 41: 687-708. 10.1016/S0735-1097(02)02885-1View ArticlePubMedGoogle Scholar
- Schiller NB, Shah PM, Crawford M, DeMaria A, Devereux R, Feigenbaum H, Gutgesell H, Reichek N, Sahn D, Schnittger I, .: Recommendations for quantitation of the left ventricle by two-dimensional echocardiography. American Society of Echocardiography Committee on Standards, Subcommittee on Quantitation of Two-Dimensional Echocardiograms. J Am Soc Echocardiogr 1989, 2: 358-367.View ArticlePubMedGoogle Scholar
- Quinones MA, Otto CM, Stoddard M, Waggoner A, Zoghbi WA: Recommendations for quantification of Doppler echocardiography: a report from the Doppler Quantification Task Force of the Nomenclature and Standards Committee of the American Society of Echocardiography. J Am Soc Echocardiogr 2002, 15: 167-184. 10.1067/mje.2002.120202View ArticlePubMedGoogle Scholar
- Zoghbi WA, Enriquez-Sarano M, Foster E, Grayburn PA, Kraft CD, Levine RA, Nihoyannopoulos P, Otto CM, Quinones MA, Rakowski H, Stewart WJ, Waggoner A, Weissman NJ: Recommendations for evaluation of the severity of native valvular regurgitation with two-dimensional and Doppler echocardiography. J Am Soc Echocardiogr 2003, 16: 777-802. 10.1016/S0894-7317(03)00335-3View ArticlePubMedGoogle Scholar
- Spencer KT, Bednarz J, Rafter PG, Korcarz C, Lang RM: Use of harmonic imaging without echocardiographic contrast to improve two-dimensional image quality. Am J Cardiol 1998, 82: 794-799. 10.1016/S0002-9149(98)00457-3View ArticlePubMedGoogle Scholar
- Kornbluth M, Liang DH, Paloma A, Schnittger I: Native tissue harmonic imaging improves endocardial border definition and visualization of cardiac structures. J Am Soc Echocardiogr 1998, 11: 693-701.View ArticlePubMedGoogle Scholar
- Atar S, Feldman A, Darawshe A, Siegel RJ, Rosenfeld T: Utility and diagnostic accuracy of hand-carried ultrasound for emergency room evaluation of chest pain. Am J Cardiol 2004, 94: 408-409. 10.1016/j.amjcard.2004.04.052View ArticlePubMedGoogle Scholar
- Vignon P, Frank MB, Lesage J, Mucke F, Francois B, Normand S, Bonnivard M, Clavel M, Gastinne H: Hand-held echocardiography with Doppler capability for the assessment of critically-ill patients: is it reliable? Intensive Care Med 2004, 30: 718-723. 10.1007/s00134-003-2128-xView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.