A new ultrasonic process for a renewal of aortic valve decalcification
Cardiovascular Ultrasound volume 4, Article number: 2 (2006)
Aortic valve decalcification by ultrasound was given up. We evaluated a new ultrasound microhandpiece (Dissectron Penstyle®) to rehabilitate this alternative treatment.
We used under magnifying lenses the ultrasound microhandpiece to decalcify 30 explanted aortic valves. In the cases with embedded calcifications the thin top of the probe could be introduced into the thickness of the leaflet preserving covering layers.
The leaflets were totally decalcified and flexible, and surrounding structures were preserved as assessed by histological examination.
This new approach of ultrasonic aortic valve decalcification gives good in vitro results which allow to consider a clinical evaluation of this procedure.
The aortic valve debridement by ultrasound in degenerative-calcific aortic stenosis appeared to be an alternative treatment for severe calcified aortic stenosis , but was given up because of the high incidence of restenosis and aortic regurgitation [2, 3]. The aim of this study was to rehabilitate aortic valve decalcification by a new probe: the microhandpiece Dissectron Penstyle®. This probe is more handy and more precise than the previous ones. Therefore the reintroduction of valve decalcification by ultrasound is envisageable.
We used ultrasonic energy to decalcify 30 aortic human tricuspid valves after they had been surgically removed during an aortic valve replacement. The valves had not been selected and were severely calcified in accordance with the necessity of the valve replacement. Ultrasonic decalcification was performed with the Dissectron® Penstyle (Integra NeuroSsciences, Sophia Antipolis, France, EU) used in neuro and general surgery but never used in cardiac surgery. This ultrasonic system consists of a control console with a handheld aspirator that contains a magnetostrictive transducer which converts electrical energy into mechanical motion (piezoelectric technology). The handpiece (Figure 1) has a hollow titanium tip (microsonotrode) that vibrates longitudinally at 30 kHz (comparing with 30 to 60 Hz used in diagnostic echocardiography), thereby fragmenting tissue in contact with its tip proportionally to the water content of the tissue. The end diameter of the Penstyle® was 1.45 mm. The excursion amplitude of the tip could be adjusted from 77 to 154 μm and was set arbitrarily at 110 μm. A side port provided continuous saline solution irrigation and the suspended particles were aspirated through the hollow tip. The irrigation was adjusted to 3 cm3/min and the suction on top of the Dissectron® was adjusted to 10–15 mmHg. The irrigating fluid avoided the overheating of the probe. The effects on tissue are the result of the combination of three parameters: ultrasonic energy, irrigation and suction. The ultrasound probe was placed in direct contact with the calcium and the generator was activated by depressing a foot pedal. We used magnifying lenses (3,5 X) to perform the process and the operating surgeon determined the adequacy of decalcification by inspection and palpation of the leaflets. A preliminary study on 10 other valves had allowed us to familiarise with the device and to adjust the settings. The completeness of the leaflet decalcification was studied by palpation and by histopathological examination with the help of a nonparametric grading system (from 1+ to 4+).
The Dissectron® handpiece was very ergonomic. The calcium was disintegrated in about 15 minutes in all the 30 cases. At the end of the process all the leaflets were observed as flexible again by palpation assessment. In 26 cases leaflets were totally decalcified (4+) (Figure 2) and in 4 cases almost totally decalcified (3+) (Figure 3). At an amplitude setting of 0.5 (about 110 μm), calcified deposits on the aortic cusps could be easily fragmented and removed, thereby restoring leaflet mobility. It was easier to decalcify aortic valve with calcific lumps only on the aortic surface of the cusps (n = 6). The preliminary study showed us that the optimal angle between the top of the Dissectron® and the valve surface is about 45°. With an angle of 45° we did not provide any perforation in the 30 valves. In the cases with embedded calcification (n = 24) we performed the decalcification by introducing the probe through the aortic surface and directing it inside the thickness of the valve in contact with the calcification (Figure. 4). On microscopic examination, sections of aortic leaflets presented a cavity instead of the calcification and preserved surrounding structures (Figure. 5).
The penstyle handpiece was totally adapted to decalcify aortic valves quickly, with comfort and precision. Although the initial results of the ultrasound decalcification with the previous probes in aortic stenosis were impressive [4, 5] two principal problems were observed: early occurrence of aortic insufficiency and significant incidence of restenosis [2, 3]. Aortic insufficiency was caused by leaflet retraction and the loss of central coaptation thought to be secondary to the healing response . The aortic restenosis was caused by the accumulation of calcium in the remaining fibrillar structure . The excursion of our tip was 110 μm comparatively to the 154 μm excursion of the Cavitron® used by McBride . As the Penstyle® handpiece decalcifies in better conditions than those previously described , with a lower level of mechanical energy, we expected a decreased healing response and consequently a decreased occurrence of aortic insufficiency. There is not any biological effects of our ultrasounds on the normal tissue, therefore we can use our probe safely in a therapeutic approach. The 1.54 mm end diameter of the Penstyle® was extremely precise to decalcify, rendering subsequent damaging of the normal valvular tissue around the calcium limited. The thin top of the probe allowed decalcification within the thickness of the leaflets without destroying the covering layers. The use of magnifying lenses facilitated an accurate procedure. This new approach of decalcification damages the valve as little as possible, so we hope to decrease the risk of early recalcification of the remaining valve. The ultimate test is the longevity of the improved flexibility and function in the patient, so we plan a clinical trial in patients to verify these two parameters. Lastly, we must distinguish two groups: calcific lumps on the aortic surface of the cusps favourable for ultrasound decalcification and embedded calcifications requiring an indispensable complementary treatment of the remaining cavities predisposing to early recalcification. The proposed indication for this new approach is a coronary bypass grafting associated with a stenosis of the aortic valve, when we hesitate to replace the valve. The expected advantage is to avoid a further operation to replace the aortic valve, with all the risks of a redo-operation.
This new approach of ultrasonic aortic valve decalcification was efficient to decalcify all the valves whatever the degree of calcification. Calcific lumps on the aortic surface of the cusps are favourable for this treatment. The mainly pitfall of our technique is to avoid to decalcify embedded calcifications. Even if it is possible to decalcify this sort of calcification, the remaining cavity will be recalcified very quickly. Only a clinical evaluation of this procedure will demonstrate a decreased risk of both recalcification and secondary aortic regurgitation.
Enright LP, Hancock EW, Shumway NE: Aortic debridment:long-term follow-up. Circulation 1971, 43: 68-72.
Craver JM: Aortic valve debridement by ultrasonic surgical aspirator: a word of caution. Ann Thorac Surg 1990, 49: 746-52.
Cosgrove DM, Ratliff NB, Schaff HV, Edwards WD: Aortic valve decalcification: history repeated with a new result. Ann Thorac Surg 1990, 49: 689-90.
Kellner HJ, Pracki P, Hildebrandt A, Binner C, Eisele G, Struck E: Aortic valve debridement by ultrasonic surgical aspirator in degenerative, aortic valve stenosis: follow-up with Doppler echocardiography. Eur J Cardiothorac Surg 1996, 10: 498-504. 10.1016/S1010-7940(96)80414-3
Baeza OR, Majid NK, Conroy DP, Donahoo JS: Combined conventional mechanical and ultrasonic debridement for aortic valvular stenosis. Ann Thorac Surg 1992, 54: 62-7.
Dahm M, Dohmen G, Groh E, Groh E, Krummenauer F, Hafner G, Mayer E, Hake U, Oelert H: Decalcification of the aortic valve does not prevent early recalcification. J Heart Valve Dis 2000, 9: 21-26.
McBride LR, Naunheim KS, Fiore AC, Harris HH, Willman VL, Kaiser JC, Pennington DG, Labovitz AJ, Barner HB: Aortic valve decalcification. J Thorac Cardiovasc Surg 1990, 100: 36-42.
Eguaras MG, Saceda JL, Luque I, Concha M: Mitral and aortic valve decalcification by ultrasonic energy. Experimental report. J Thorac Cardiovasc Surg 1988, 95: 1038-40.
The author(s) declare that they have no competing interests.
SA did all the research and wrote the paper.
EV had been involved in the research.
LC had been involved in the anatomopathology examination.
FF had been involved in the analysis of the data.
OJ had been involved in the final approval of the version to be published.
About this article
Cite this article
Aubert, S., Voiglio, E., Chalabreysse, L. et al. A new ultrasonic process for a renewal of aortic valve decalcification. Cardiovasc Ultrasound 4, 2 (2006). https://doi.org/10.1186/1476-7120-4-2