This article has Open Peer Review reports available.
Head to head comparison of 2D vs real time 3D dipyridamole stress echocardiography
© Varnero et al; licensee BioMed Central Ltd. 2008
Received: 15 May 2008
Accepted: 20 June 2008
Published: 20 June 2008
Real-time three-dimensional (RT-3D) echocardiography has entered the clinical practice but true incremental value over standard two-dimensional echocardiography (2D) remains uncertain when applied to stress echo. The aim of the present study is to establish the additional value of RT-3D stress echo over standard 2D stress echocardiography. We evaluated 23 consecutive patients (age = 65 ± 10 years, 16 men) referred for dipyridamole stress echocardiography with Sonos 7500 (Philips Medical Systems, Palo, Alto, CA) equipped with a phased – array 1.6–2.5 MHz probe with second harmonic capability for 2D imaging and a 2–4 MHz matrix-phased array transducer producing 60 × 70 volumetric pyramidal data containing the entire left ventricle for RT-3D imaging. In all patients, images were digitally stored in 2D and 3D for baseline and peak stress with a delay between acquisitions of less than 60 seconds. Wall motion analysis was interpreted on-line for 2D and off-line for RT-3D by joint reading of two expert stress ecocardiographist. Segmental image quality was scored from 1 = excellent to 5 = uninterpretable. Interpretable images were obtained in all patients. Acquisition time for 2D images was 67 ± 21 sec vs 40 ± 22 sec for RT-3D (p = 0.5). Wall motion analysis time was 2.8 ± 0.5 min for 2D and 13 ± 7 min for 3D (p = 0.0001). Segmental image quality score was 1.4 ± 0.5 for 2D and 2.6 ± 0.7 for 3D (p = 0.0001). Positive test results was found in 5/23 patients. 2D and RT-3D were in agreement in 3 out of these 5 positive exams. Overall stress result (positive vs negative) concordance was 91% (Kappa = 0.80) between 2D and RT-3D. During dipyridamole stress echocardiography RT-3D imaging is highly feasible and shows a high concordance rate with standard 2D stress echo. 2D images take longer time to acquire and RT-3D is more time-consuming to analyze. At present, there is no clear clinical advantage justifying routine RT-3D stress echocardiography use.
Two-dimensional dipyridamole stress echocardiography, is an established and validated method for both the diagnosis and prognosis [1–5] of patients with known or suspected coronary artery disease. However grounds for an accurate interpretation in stress echo rest on two important features: first-acoustic windows that permits complete endocardial border visualization within proper planes of left ventricle (LV) and secondly-prompt acquisition of peak images pertaining predictive accuracy. Unfortunately, attempting to image a three-dimensional structure, such as the heart, by a 2D device requires multiple windows and time, with the danger of still producing insufficient tomographic view and losing important information between section planes . Real-time three-dimensional (RT-3D) echocardiography has entered the clinical practice  but true incremental value over standard two-dimensional echocardiography (2D) remains uncertain in stress echo. The three-dimensional echocardiography (3D) has the theoretical potential to more completely assess LV [8–12] but previous 3D imaging system were tedious techniques that used off-line reconstruction of multiple 2D images [8–10], not suitable for stress echocardiography. RT-3D, a much more user-friendly technique, now permits single-window and single-heart beat acquisition of complete LV segments in a volume-shaped cineloop , having the prerequisite to employed during stress echo. The aim of the present study was to assess the additional value of RT3D over conventional 2D dipyridamole stress echocardiography.
The study population consisted of 23 consecutive patients (age = 65 ± 10 years, 16 men) with known or suspected coronary artery disease referred for clinically indicated stress echocardiography. Patients were prospectively enrolled in Ospedale di Savona, Italy with the following criteria: age ≥ 18 years; adequate echocardiogram to assess regional wall motion in 2D and RT-3D (the echocardiogram was considered adequate if ≥ 13 of the 17 segments were visualized). Exclusion criteria included: poor acoustic window, contraindications to dipyridamole, recent (< 1 month) episode of ventricular fibrillation, significant stenosis of the aortic valve, and patient refusal to enter the study. Decisions concerning medical therapy at the time of testing and/or coronary angiography were left to the attending physician. A quantitative luminal narrowing ≥ 50% was considered significant. All patients gave informed written consent prior to dipyridamole stress echocardiography.
Two-dimensional echocardiography images were performed with Sonos 7500 (Phillips Medical Systems, Palo, Alto, CA) equipped with a phased array 1.6–2.5 MHz probe with second harmonic capability. In all patients, the four standard views (parasternal long and short axis, apical 4 and 2 chambers) were obtained at baseline and peak stress and were recorded on super-VHS and digitally stored.
Real-time 3D imaging
Patients were asked to abstain from food and drinks containing xanthine for ≥ 24 hrs prior to the study. All patients performed a dipyridamole stress echo. Dipyridamole was given at a maximal dose of 0.84 mg/kg in 10 minutes, unless symptoms of intolerability, positivity or hypotension (relative or absolute, >30 mmHg decrease in blood pressure) occurred. Test was considered positive for ischemia if any new or worsening dyssynergy occurred in more than 1 contiguous segment of the same vascular territory. A segment was considered to be viable when it improved by one grade or more at peak stress (for instance, a hypokinetic segment becoming normal, or an akinetic segment becoming hypokinetic). The left ventricle was divided into 17 segments as suggested the American heart Association . Segmental wall motion was graded as follow: normal = 1, hypokinetic = 2, akinetic = 3, and dyskinetic = 4. Wall motion score index was derived by dividing the sum of individual visualized segment scores by the number of visualized segments. It has been agreed a priori to grade as normal any "mild" or "questionable" hypokinesia. Segmental image quality was assessed as previously reported  briefly a score of 1 = complete endocardial definition and wall thickening both at rest and at peak, 2 = visualization of all segments but not as well as , 3 = inadequate visualization ≤ 2 segments but adequate adjacent segments of same territory, 4 = inadequate visualization ≥ 3 segments but adequate adjacent segments of same territory, 5 = inadequate visualization comprising 1 or whole territories. Segmental analysis (wall motion and image quality) was done by joint reading of two expert observers on-line for 2D stress echo and off-line for RT-3D.
Variables are expressed as mean values ± SD. Student paired-sample t test was used to compare continuous variables. Cohen's coefficient of variation, Kappa was used to assess agreement between 2D and RT-3D results. A Kappa value ≥ 0.45 was considered to be a good agreement while Kappa ≥ 0.75 was considered to be an excellent agreement. A p value < 0.05 was considered significant. All calculations were made using SPSS software (SPSS version 13.0 for windows, 1995).
n = 23 pts
65 ± 10
Calcium channel blockers
Feasibility of RT-3D echocardiography
Concordance 2D vs 3D: per patient analysis
Real-time three-dimensional dipyridamole stress echocardiography is highly feasible in clinical practice. RT-3D allowed the rapid (~40 sec) acquisition at peak stress of a volumetric cineloop that contained the entire LV and to review multiple, standard 2D images. The inferior image quality of RT3D might partly be explained by the round and flat face of the transducer which offers poor intercostals skin contact and the low frequencies used . Moreover the frame rates decrease further with an increase of heart rate (i.e.: during dobutamine infusion or during exercise) with consequent reduction of spatial resolution. In these study we used the dipyridamole 0.84 mg/kg in 10 minutes and the hemodynamic response to this drug is different in term of increase of heart frequency in comparison to dobutamine or exercise stress echo. For this reason the dypiridamole echo test could be the best choice in patients submitted to RT3D stress echo. In this study we showed a good to excellent concordance with 2D echo on wall motion analysis for detection of inducible ischemia.
Comparison with previous studies
Previous studies on three-dimensional imaging used a different technology than real-time three-dimensional echocardiography, these previous techniques were dependant on both ECG and respiratory gating because 3D images were a composite of multiples 2D acquisitions and cardiac cycles, thus highly affected by translation motion, arrhythmia and improper spatial registration . Due to these limitations they never entered the clinical practice.
Real-time three-dimensional echocardiography has the prerequisites to analyze segmental wall motion, volume, etc. In fact, it has been demonstrated that [6, 13–18] using real-time technology it was possible to image, more rapidly, most of LV segments by a single-window acquisition: 93–98% of LV segments from parasternal volume set and 85–89% from apical volume set [6, 14]. It was also demonstrated that RT-3D had a high concordance (Kappa = 0.52–0.72) with 2D reading . In our study, we achieved similar rates of LV segments visualization but there was only a non significant trend toward a shorter time of acquisition with RT-3D when compared with 2D which might be accounted for an appropriate learning curve time. The time necessary to RT3D imaging optimization was measured and it may have prolonged significantly the acquisition time.
RT3D stress echocardiography has been previously shown to be highly feasible for both dobutamine [14–18] and exercise  stress testing but the true incremental value was never proven and only one study showed higher sensitivity than 2D imaging (87.9% vs 79.3%) . In the present we did not address the diagnostic accuracy of RT-3D vs 2D echo but other authors have demonstrated comparable sensitivity (93%), specificity (75%), and accuracy (99%) between the two methods .
RT3D has entered the clinical arena but no additional value over conventional 2D echocardiography during stress echocardiography can be demonstrated. Recent ASE and EAE consensus statements on stress echocardiography did not recommend the routine use of this technology during stress echocardiography even though it may shorten significantly time of acquisition counterbalanced by a longer time of data-set analysis. Nonetheless RT3D may play a role in stress echocardiography when wall motion analysis is not the target. The assessment of contractility or the function of the right ventricle can become the perfect clinical setting for this technique. More studies are warranted to assess its real clinical value in different subsets of patients but the evidence is sound enough to state that RT3D is here to stay.
- Severi S, Picano E, Michelassi C, Lattanzi F, Landi P, Distante A, L'Abbate A: Diagnostic and prognostic value of dipyridamole echocardiography in patients with suspected coronary artery disease. Comparison with exercise electrocardiography. Circulation. 1994, 89: 1160-1173.View ArticlePubMedGoogle Scholar
- Picano E, Bedetti G, Varga A, Cseh E: The comparable diagnostic accuracies of dobutamine-stress and dipyridamole-stress echocardiography: a meta-analysis. Coron Artery Dis. 2000, 11 (2): 151-9. 10.1097/00019501-200003000-00010.View ArticlePubMedGoogle Scholar
- Picano E, Landi P, Bolognese L, Chiarandà G, Chiarella F, Seveso G, Sclavo MG, Gandolfo N, Previtali M, Orlandini A: Prognostic value of dipyridamole echocardiography early after uncomplicated myocardial infarction: a large-scale, multicenter trial. Am J Med. 1993, 95 (6): 606-18. 10.1016/0002-9343(93)90357-U.View ArticleGoogle Scholar
- Sicari R, Nihoyannopoulos P, Evangelista A, Kasprzak J, Lancellotti P, Poldermans D, Voigt JU, Zamorano JL, on behalf of the European Association of Echocardiography: Expert consensus statement on stress echocardiography from the European Association of Echocardiography. Eur J Echocardiogr. 2008Google Scholar
- Picano E: Stress echocardiography. 2003, Springler-Verlag Berlin, Heilderberg, New York, 4View ArticleGoogle Scholar
- Collins M, Hsieh A, Ohazama CJ, Ota T, Stetten G, Donovan CL, Kisslo J, Ryan T: Assessment of regional wall motion abnormalities with real-time 3-dimensional echocardiography. J Am Soc Echocardiogr. 1999, 12: 7-14. 10.1016/S0894-7317(99)70167-7.View ArticlePubMedGoogle Scholar
- Picard M: The time for 3D. J Am Soc Echocardiogr. 2007, 20: 19A-20A.View ArticlePubMedGoogle Scholar
- Maehle J, Bjoernstad K, Aakhus S, Torp HG, Angelsen BA: Three-dimensional echocardiography for quantitative left ventricular wall motion analysis: a method for reconstruction of endocardial surface and evaluation of regional dysfunction. Echocardiography. 1994, 11: 397-408. 10.1111/j.1540-8175.1994.tb01382.x.View ArticlePubMedGoogle Scholar
- Raichlen JS, Trivedi SS, Herman GT, St John Sutton MG, Reichek N: Dynamic 3-dimensional reconstruction of the left ventricle from 2-dimensional echocardiograms. J Am Coll Cardiol. 1986, 8: 364-70.View ArticlePubMedGoogle Scholar
- Lange A, Przemyslaw P, Burstow DJ, Godman MJ: Three-dimensional echocardiography: historical development and current applications. J Am Soc Echocardiogr. 2001, 14: 403-12. 10.1067/mje.2001.113147.View ArticlePubMedGoogle Scholar
- Cerqueira MD, Weissman NJ, Dilsizian V, Jacobs AK, Kaul S, Laskey WK, Pennell DJ, Rumberger JA, Ryan T, Verani MS, American Heart Association Writing Group on Myocardial Segmentation and Registration for Cardiac Imaging: merican Heart Association Writing Group on Myocardial Segmentation and Registration for Cardiac Imaging. Standardized myocardial segmentation and nomenclature for tomographic imaging of the heart: a statement for healthcare professionals from the Cardiac Imaging Committee of the Council on Clinical cardiology of the American Heart Association. Circulation. 2002, 105: 539-42. 10.1161/hc0402.102975.View ArticlePubMedGoogle Scholar
- Hoffmann R, Lethen H, Marwick T, Arnese M, Fioretti P, Pingitore A, Picano E, Buck T, Erbel R, Flachskampf FA, Hanrath P: Analysis of interinstitutional observer agreement in interpretation of dobutamine stress echocardiograms. J Am Coll Cardiol. 1996, 27: 330-6. 10.1016/0735-1097(95)00483-1.View ArticlePubMedGoogle Scholar
- Takuma S, Zwas DR, Fard A, Wu H, Chaudhry H, Di Tullio MR, Ota T, Homma S: Real-time, 3-dimensional echocardiography acquires all standard 2-dimensional images from 2 volume sets: a clinical demonstration in 45 patients. J Am Soc Echocardiogr. 1999, 12: 1-6. 10.1016/S0894-7317(99)70166-5.View ArticlePubMedGoogle Scholar
- Ahmad M, Xie T, McCulloch M, Abreo G, Runge M: Real-time three-dimensional dobutamine stress echocardiography in assessment of ischemia: comparison with two-dimensional dobutamine stress echocardiography. J Am Coll Cardiol. 2001, 37: 1303-9. 10.1016/S0735-1097(01)01159-7.View ArticlePubMedGoogle Scholar
- Matsumura Y, Hozumi T, Arai K, Sugioka K, Ujino K, Takemoto Y, Yamagishi H, Yoshiyama M, Yoshikawa J: Non-invasive assessment of myocardial ischaemia using new real-time three-dimensional dobutamine stress echocardiography: comparison with conventional two-dimensional methods. Eur Heart J. 2005, 26 (16): 1625-32. 10.1093/eurheartj/ehi194.View ArticlePubMedGoogle Scholar
- Eroglu E, D'hooge J, Herbots L, Thijs D, Dubois C, Sinnaeve P, Dens J, Vanhaecke J, Rademakers F: Comparison of real-time tri-plane and conventional 2D dobutamine stress echocardiography for the assessment of coronary artery disease. Eur Heart J. 2006, 27 (14): 1719-24. 10.1093/eurheartj/ehl023.View ArticlePubMedGoogle Scholar
- Takeuchi M, Otani S, Weinert L, Spencer KT, Lang RM: Comparison of contrast-enhanced real-time live 3-dimensional dobutamine stress echocardiography with contrast 2-dimensional echocardiography for detecting stress-induced wall-motion abnormalities. J Am Soc Echocardiogr. 2006, 19: 294-9. 10.1016/j.echo.2005.10.008.View ArticlePubMedGoogle Scholar
- Pulerwitz T, Hirata K, Abe Y, Otsuka R, Herz S, Okajima K, Jin Z, Di Tullio MR, Homma S: Feasibility of using a real-time 3-dimensional technique for contrast dobutamine stress echocardiography. J Am Soc Echocardiogr. 2006, 19 (5): 540-5. 10.1016/j.echo.2005.12.006.View ArticlePubMedGoogle Scholar
- Zwas DR, Takuma S, Mullis-Jansson S, Fard A, Chaudhry H, Wu H, Di Tullio MR, Homma S: Feasibility of real-time 3-dimensional treadmill stress echocardiography. J Am Soc Echocardiogr. 1999, 12: 285-9. 10.1016/S0894-7317(99)70048-9.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.