Research | Open | Open Peer Review | Published:
Spatial orientation of cross-sectional images of coronary arteries: point of view in intracoronary imaging
Cardiovascular Ultrasoundvolume 10, Article number: 12 (2012)
In studies where cross-sectional images of coronary arteries obtained with different imaging modalities are compared, the importance of correct co-localization and matching of images along the coronary artery longitudinal axis is obvious. However, it appears neglected that correct spatial orientation of the cross-sectional plane may not be obtainable just by rotating the images to ensure co-localization of identifiable landmarks such as sidebranches. A cross-section has two sides, one facing proximally and the other distally, and pairs of images reconstructed corresponding to these opposite points of view are mirror images of each other and not superimposable. This may be difficult if not impossible to recognize and unrecognized it will give rise to flawed results in the development and validation of imaging technologies aimed at plaque characterization (tissue mapping). We determined the imagined point of view for three commercially available intracoronary imaging systems used by invasive cardiologists and illustrate its importance in imaging modality validation.
Methods and Results
We made an asymmetric phantom and investigated it with two different intravascular ultrasound (IVUS) systems and one optical coherence tomography (OCT) system. The asymmetry of the phantom allowed determination of the spatial orientation of the cross-sectional images. On all tested systems, an observer should imagine herself/himself standing proximal to the cross-section when looking at the intravascular images.
The tested intracoronary imaging modalities displayed cross-sectional images with a spatial orientation corresponding to a proximal point of view. Knowledge of the spatial orientation is mandatory when comparing and validating different imaging modalities aimed at plaque characterization.
Many intravascular ultrasound (IVUS) and optical coherence tomography (OCT) cross-sectional images of coronary arteries are published in biomedical journals. Sometimes these images are presented together with images from microscopic examination of the same arterial positions illustrating how the imaging modality correctly displays tissue characteristics [1, 2].
When comparing cross-sectional images obtained with different techniques, such as intravascular imaging and microscopy, efforts must be made to assure that the images are from exactly the same arterial position along the longitudinal axis of the coronary artery, and this is often devoted a lot of attention.
Also, when comparing cross-sectional images, rotational adjustment is usually required to obtain correct co-localization of identifiable landmark structures such as sidebranches. Rotation of a cross-sectional image does not change area measurements or intraplaque location of a certain component of interest. Therefore rotation does not affect interpretation.
However, in addition to longitudinal position and rotation, cross-sectional images also possess a spatial orientation of the cross-sectional plane. This is only rarely commented on but cannot be taken as lightly as rotation [3, 4]. Changing spatial orientation, and thereby our imagined point of view, changes intraplaque location of plaque components in the cross-sectional plane. Thereby spatial orientation has implications in studies focusing on the identification and localization of plaque components (tissue mapping).
We determined the spatial orientation of cross-sectional images displayed on two different IVUS systems and one OCT system and exemplify implications.
Spatial orientation of microscopic sections
When a section is cut for microscopic examination, the section may be laid down on the glass slide on one of two sides. Photomicrographs of a section lying on different sides will be mirror images of each other. So, although they are photomicrographs of the same section, i.e. similar position and rotation, the intraplaque localizations of plaque components will differ (Figure 1).
With strict adherence to procedures throughout the preparation of arterial sections for microscopic examination, the same known spatial orientation of sections can be obtained. Thereby, with proper attention to orientation in the preparation of tissue section for microscopy, one can always decide and know the imagined point of view, i.e., whether one should imagine looking at a coronary artery cross-section standing proximal (at the coronary ostium) or distal to the cross section (Figure 1).
Spatial orientation of intravascular images
An intravascular imaging catheter looks from the arterial lumen and outward on the arterial wall. But the displayed images created with intravascular imaging represent cross-sections that have one of two possible spatial orientations, just like the microscopic section. Throughout the pull back, the observer should imagine herself/himself standing either proximal or distal to the cross-section when looking at the cross-sectional images (Figure 1). The spatial orientation and thereby the imagined point of view is decided by the reconstruction algorithm. When making comparisons of intravascular and microscopic images, one has to know the spatial orientation of the intravascular images in addition to the images from microscopy and make sure they are the same before any comparison is meaningful. To know the spatial orientation, one has to either know the reconstruction algorithm or determine the spatial orientation experimentally.
Revealing point of view
To determine the spatial orientation experimentally, we made an asymmetric phantom using a short piece of tubing with two wire points not directly across from each other on the cross-section of the tube. One wire alone constituted one wirepoint and two wires close together constituted the other. We then decided which end was to be considered the proximal and investigated the tube with two different IVUS systems and one OCT system inserting the IVUS and OCT catheters from the imagined proximal end (Figure 2).
The IVUS systems were Volcano In-Vision Gold system with an Eagle Eye Gold catheter and Boston Scientific iLab system with an Atlantis SR Pro catheter. The IVUS investigations were performed with the phantom submerged in water. Manual pull back recordings were used. The OCT system was Light Lab C7XR with a Lightlab C7 Dragonfly Imaging Catheter.
Results and discussion
The spatial orientation (point of view)
On the IVUS and OCT images, all three wires were clearly visible making the two asymmetrically placed wire points clearly distinguishable. Owing to the asymmety of the phantom, it was evident that the intravascular images were displayed to the observer, so that the observer should imagine herself/himself looking at the cross-sectional images standing proximal to the displayed cross-sectional image of the artery (Figure 3).
In studies comparing cross sectional images from these imaging systems to images from microscopy, investigators should then ensure that their cross sectional images from microscopy display the proximal view (same spatial orientation). Otherwise, comparison will not be meaningful.
When does knowledge about spatial orientation not make a difference?
Clinically, treatments, such as angioplasty or stenting, are applied equally to the entire arterial circumference. Therefore, for making the clinical decision whether to apply a treatment or not, the spatial orientation of the cross-sectional image does not make a difference.
In clinical research, area measurements on intravascular images are used as study end points . Measurements on areas, e.g. of a plaque or necrotic core from reconstructed images, yield the same results regardless of the spatial orientation and therefore the thin-cap fibroatheroma (TCFA)[6, 7] can be correctly identified from both points of view (Figure 1).
However, meaningful use of an intravascular imaging modality, both clinically and in clinical research, provides that the intravascular imaging modality has been properly validated for identification and quantification of the plaque components that are of interest clinically and in clinical research. I.e., the use of these imaging modalities clinically and in clinical research is dependent upon reliable validation of the imaging systems. It is in this validation that spatial orientation is important.
When does knowledge about spatial orientation make a difference?
For validation of imaging modalities focusing on the identification and quantification of plaque components, intraplaque localization is used and the spatial orientation of the cross-sectional images becomes critical (Figure 4). When the images from intravascular imaging and microscopy have the same spatial orientation, comparison makes sense. But if the images from intravascular imaging and microscopy have different spatial orientations, the compared images are not superimposable and direct comparison makes no sense. The spatial orientation can and must be known before any comparison of the images because knowledge about spatial orientation is crucial for correct interpretation of cross-sectional images and validation of technologies creating such images (Figure 4).
These considerations apply to the comparison of asymmetric cross-sectional images in general, irrespective of imaging modalities. In this study, other imaging modalities such as computed tomography, magnetic resonance imaging, near infrared spectroscopy etc., were not investigated; however the considerations about spatial orientation also apply whenever images from these modalities are compared to images from other imaging modalities, e.g. microscopy.
Intravascular imaging catheter rotation
Intravascular imaging catheters can rotate along their longitudinal axis during pull back. Therefore, spatial orientation cannot be determined based on localization of two structures present on different images from the same pull back. Neither can spatial orientation be determined based on the circumferential localization of a single landmark, such as a sidebranch, as catheter rotation randomly determines the localization of this landmark. Determining spatial orientation experimentally requires asymmetry in one image.
The tested intracoronary imaging modalities displayed cross-sectional images with a spatial orientation corresponding to an imagined point of view proximal to the cross-section. Knowledge of the point of view is mandatory when comparing and validating different imaging modalities. These considerations have implications for the development and evaluation of imaging technologies. Spatial orientation of images might appropriately be disclosed by manufacturers and discussed in consensus papers on intravascular imaging [8, 9].
Optical coherence tomography
Nair A, Kuban BD, Tuzcu EM, Schoenhagen P, Nissen SE, Vince DG: Coronary plaque classification with intravascular ultrasound radiofrequency data analysis. Circulation. 2002, 106: 2200-2206. 10.1161/01.CIR.0000035654.18341.5E
Nair A, Margolis M, Kuban B, Vince D: Automated coronary plaque characterisation with intravascular ultrasound backscatter: ex vivo validation. EuroIntervention. 2007, 3: 113-120.
Thim T, Hagensen MK, Wallace-Bradley D, Granada JF, Kaluza GL, Drouet L, Paaske WP, Botker HE, Falk E: Unreliable assessment of necrotic core by virtual histology intravascular ultrasound in porcine coronary artery disease. Circ Cardiovasc Imaging. 2010, 3: 384-391. 10.1161/CIRCIMAGING.109.919357
Thim T: Human-like atherosclerosis in minipigs: a new model for detection and treatment of vulnerable plaques. Dan Med Bull. 2010, 57: B4161-
Serruys PW, Garcia-Garcia HM, Buszman P, Erne P, Verheye S, Aschermann M, Duckers H, Bleie O, Dudek D, Botker HE, et al.: Effects of the direct lipoprotein-associated phospholipase A(2) inhibitor darapladib on human coronary atherosclerotic plaque. Circulation. 2008, 118: 1172-1182. 10.1161/CIRCULATIONAHA.108.771899
Thim T, Hagensen MK, Bentzon JF, Falk E: From vulnerable plaque to atherothrombosis. J Intern Med. 2008, 263: 506-516. 10.1111/j.1365-2796.2008.01947.x
Thim T, Hagensen MK, Drouet L, Bal dit SC, Bonneau M, Granada JF, Nielsen LB, Paaske WP, Botker HE, Falk E: Familial hypercholesterolaemic downsized pig with human-like coronary atherosclerosis: a model for preclinical studies. EuroIntervention. 2010, 6: 261-268. 10.4244/EIJV6I2A42
Mintz GS, Nissen SE, Anderson WD, Bailey SR, Erbel R, Fitzgerald PJ, Pinto FJ, Rosenfield K, Siegel RJ, Tuzcu EM, et al.: American College of Cardiology Clinical Expert Consensus Document on Standards for Acquisition, Measurement and Reporting of Intravascular Ultrasound Studies (IVUS). A report of the American College of Cardiology Task Force on Clinical Expert Consensus Documents. J Am Coll Cardiol. 2001, 37: 1478-1492. 10.1016/S0735-1097(01)01175-5
Prati F, Regar E, Mintz GS, Arbustini E, Di MC, Jang IK, Akasaka T, Costa M, Guagliumi G, Grube E, et al.: Expert review document on methodology, terminology, and clinical applications of optical coherence tomography: physical principles, methodology of image acquisition, and clinical application for assessment of coronary arteries and atherosclerosis. Eur Heart J. 2010, 31: 401-415. 10.1093/eurheartj/ehp433
The authors declare that they have no competing interests.
TT: conception and design, acquisition of data, and analysis and interpretation of data, drafting of the manuscript. EF: analysis and interpretation of data, revision of the manuscript. Both authors read and approved the final manuscript.