- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
Volume quantification by contrast-enhanced ultrasound: an in-vitro comparison with true volumes and thermodilution
© Herold et al.; licensee BioMed Central Ltd. 2013
Received: 11 July 2013
Accepted: 14 October 2013
Published: 17 October 2013
Contrast-enhanced ultrasound (CEUS) has recently been proposed as a minimally- invasive, alternative method for blood volume measurement. This study aims at comparing the accuracy of CEUS and the classical thermodilution techniques for volume assessment in an in-vitro set-up.
The in-vitro set-up consisted of a variable network between an inflow and outflow tube and a roller pump. The inflow and outflow tubes were insonified with an ultrasound array transducer and a thermistor was placed in each tube. Indicator dilution curves were made by injecting indicator which consisted of an ultrasound-contrast-agent diluted in ice-cold saline. Both acoustic intensity- and thermo-dilution curves were used to calculate the indicator mean transit time between the inflow and outflow tube. The volumes were derived by multiplying the estimated mean transit time by the flow rate. We compared the volumes measured by CEUS with the true volumes of the variable network and those measured by thermodilution by Bland-Altman and intraclass-correlation analysis.
The measurements by CEUS and thermodilution showed a very strong correlation (rs = 0.94) with a modest volume underestimation by CEUS of −40 ± 28 mL and an overestimation of 84 ± 62 mL by thermodilution compared with the true volumes. Both CEUS and thermodilution showed a high statistically significant correlation with the true volume (rs = 0.97 (95% CI, 0.95 - 0.98; P<0.0001) and rs = 0.96 (95% CI, 0.94 - 0.98; P<0.0001, respectively).
CEUS volume estimation provides a strong correlation with both the true volumes in-vitro and volume estimation by thermodilution. It may therefore represent an interesting alternative to the standard, invasive thermodilution technique.
Blood volume determination is a daily routine in anesthesia and intensive care practice. Most of the time, it is roughly estimated using clinical parameters such as blood pressure, heart frequency, urine output, and peripheral temperature. In sepsis, the postoperative phase and heart failure, circulating volume can be difficult to assess and in these cases classical dilution techniques are of additional value. The intrathoracic blood volume can be estimated by transthoracic thermodilution, presently one of the most widely used techniques. Its value and change in response to fluid challenge reflects the left ventricular preload and changes in preload better than more conventional measures like central venous pressure and pulmonary artery wedge pressure . However, these techniques are invasive and require catheterization of the heart  and/or large vessels , which can lead to complications.
With classical dilution techniques, a known amount of indicator is injected via a central venous line into the jugular or subclavian vein and is carried through the heart and pulmonary circulation where it is mixed and diluted. Downstream, the indicator concentration-change over time is measured at a detection site to create an indicator dilution curve (IDC) . The IDC is used to estimate the mean transit time (MTT); this is the average time it takes for the indicator to travel from the injection site to the detection site . When two detection sites are used, the product of MTT difference and flow can be used to calculate the volume in between both sites. Classical indicator-dilution techniques can be performed with different standard indicators (such as cold saline, indocyanine green or lithium) through different access sites (e.g. right or left heart-sided) [2–4, 6]. The transpulmonary thermodilution technique allows measurement of cardiac output (CO) and intrathoracic blood volumes [3, 6].
A less invasive technique may be a valuable alternative to these methods, which are hampered by their invasiveness. A promising minimally invasive alternative technique uses an ultrasound contrast agent (UCA) injected into a peripheral vein as indicator. This can be detected noninvasively by contrast-enhanced ultrasound (CEUS) imaging. Mischi et al. previously demonstrated that this technique can be used for estimating blood volumes [7–9]. In this study the measurement of blood volumes by means of UCA dilution with transthoracic echography (TTE) was tested and validated in-vitro. The determination coefficient between the real and the estimated volumes was larger than 0.999 in different model fits .
However, to date there has been no comparison with the classic thermodilution technique, which is clinically considered the gold standard for CO and blood volume measurement.
The aim of our study is to compare the CEUS with the thermodilution technique for volume quantification in an in-vitro set-up with different flows and volumes. We decided to use the transesophageal probe as this probe is often used in the perioperative setting, where large volume shifts can occur.
Ultrasound system and settings
A commercially available scanner (iE33, Philips Healthcare, Andover, MA, USA) was used to obtain cross-sectional B-mode images of the inflow and outflow tubes. Harmonic imaging at 2.7 - 5.4 MHz was used in order to increase the signal-to-noise ratio (SNR) for low UCA concentration together with a low mechanical index (MI) of 0.2 to reduce bubble disruption. Frame rate was set at 27 Hz, the same time-gain and lateral-gain compensation were employed over all measurements, compression was set at 50 dB, general gain at 60%, and image depth was 8 cm with the focus being at the level of both tubes.
Thermodilution measurements were performed using the pressure wires as described above. These pressure wires have temperature sensing tips that were positioned in the polyurethane tubes and were intercepted by the ultrasound beam for contrast quantification. The temperature sensors of both pressure wires were connected to a Wheatstone bridge adjusted to half-bridge configuration in order to output measured IDCs from both sensors. The electrical circuit further comprised a feedback amplifier (INA 118, Burr-Brown Corporation, Tucson, AZ, USA), a power supply (Delta Elektronica, Zierikzee, the Netherlands), and a data acquisition board (NI USB-6341, National Instruments, Austin, TX, USA). The bridge was balanced by manual adjustment of the value of an embedded potentiometer. The output signal was amplified in such a way that the full range of the analog-to-digital converter of the data acquisition card (0-10 V) was exploited. High frequency noise suppression was achieved by placing an additional capacitance in parallel with the input impedance of the amplifier. All devices were shielded and grounded to minimize ambient disturbances. The thermodilution curves were acquired with LabVIEW (National Instruments, Austin, TX, USA) and processed in MATLAB® 2009b (The Mathworks, Natick, MA, USA). The full system was calibrated by mapping the measured voltage as a function of temperature in a water-filled basin measured by a digital thermometer (Keithley 871, Keithley Instruments, Cleveland, OH, USA). The calibration showed a linear relationship with a slope of 0.65 V/°C and r2 = 0.999. These results confirmed the system linearity for temperatures in a range 24°C – 40°C.
Ultrasound contrast measurement
Additional file 1: Transesophageal echography view of the inflow and outflow tube in the water-filled basin of the in-vitro set-up. The change in acoustic intensity was analyzed with commercially available software (QLAB 8, Philips Healthcare, Andover, MA, USA). This software allows drawing of multiple regions of interest (ROIs) to obtain acoustic IDCs. Two ROIs were drawn within the thin polyurethane layer of the inflow and outflow tube. (MOV 10 MB)
All data were reported as mean values ± standard deviation (SD) or as median ± interquartile range (IQR) depending on the distribution of the variables of three consecutive measurements. The first goal was to investigate the agreement between measured volumes by both techniques and the true set-up volumes. Statistical significance was considered as a two-sided P<0.05. Bland-Altman analysis was used to determine the agreement between measured volumes and the true volumes . The effect of the different flows on the volume measurement was also investigated and reported in dedicated plots. Reproducibility was assessed by the intraclass-correlation coefficient (ICC). ICC consists of a basic calculation as repeated-measures analysis of variance (ANOVA) and the intraobserver reliability (ICC (1,1)). ICC assesses the agreement of quantitative variables on its reliability and consistency [14, 15]. The second goal was to analyze the correlation between the CEUS volumes and thermodilution volumes, assuming thermodilution as the gold standard. Correlation coefficients were assessed using the Pearson correlation coefficient R or the Spearman correlation coefficient rs depending on normal distribution or non-normal distribution of variables, respectively. Statistical analysis was performed using GraphPad Prism version 5.03 (GraphPad Software, San Diego, CA, USA) except for the intraclass-correlation, which was analyzed by Unistat® Statistical Package for Windows™ version 6.0 (Unistat House, London, England). Statistical analysis was performed by using all data (n=72 measurements) to exclude a bias.
A total of 79 measurements were performed. Seven measurements could not be used for analysis as a result of failed acquisition on the ultrasound equipment (n=5) or due to technical failure of the Wheatstone bridge (n=2). All remaining 72 measurements were used for analysis. The CEUS derived median volume of these, using the LDRW model double fit method was 590 (394–764) mL. For the impulse response method, the volume was 574 (382 –725) mL. The median volumes estimated with the thermodilution technique double fit method and impulse response method were 722 (489 – 944) mL and 693 (459 – 886) mL, respectively.
Repeated-measures ANOVA demonstrated no significant variance between the measures for both CEUS and thermodilution-derived volumes. Intraclass-correlation between three repetitive measurements was ICC = 0.99 (95% confidence interval (CI), 0.98 - 1.00) for the CEUS derived volumes and ICC = 0.97 (95% CI, 0.94 - 0.98) for the thermodilution calculated volumes, using the double fit method. The intraclass-correlation for the measured volumes using the impulse response method was ICC = 0.98 (95% CI, 0.96 - 0.99) for CEUS and ICC = 0.98 (95% CI, 0.97 - 0.99) for thermodilution.
Effect of flow and volume on the measurements
Correlation between the measured volumes and true volumes
Correlation between volumes estimated with CEUS and the thermodilution technique
The correlation between CEUS and thermodilution technique showed rs = 0.94 (95% CI, 0.90 - 0.96; P<0.0001), using the LDRW double fit method and rs = 0.97 (95% CI, 0.95 - 0.98; P<0.0001), using the impulse response method. The Bland-Altman analysis showed a bias of −124 ± 74 mL using the LDRW model double fit method, where thermodilution estimated volumes were on average larger than CEUS volumes. For the LDRW model impulse response method the bias was −108 ± 67 mL.
In the present study, comparing volume measurement by CEUS with thermodilution in a controlled in-vitro set-up, we have demonstrated that there is a good correlation between volumes measured by CEUS and by thermodilution (rs = 0.94 with the LDRW double fit method). Using the Bland-Altman analysis, there was a good level of agreement (bias −108 ± 67 mL with the LDRW double fit method) between both methods with only a modest underestimation of the true volume by CEUS (bias −40 ± 28 mL with the LDRW double fit method). Interestingly, compared to the gold standard of thermodilution, CEUS demonstrated to provide a more accurate measure of a known volume in this in-vitro set-up with a bias of only −40 ± 28 mL compared to 84 ± 62 mL for the thermodilution method.
In general, the thermodilution technique overestimated all volumes (Figure 4). The overestimation of the thermodilution volumes can be explained, in the first place, by loss of heat to the surroundings due to conduction across the tube wall. Heat loss is more marked at low flows and large volumes (Figure 4) due to longer contact-time and larger surface area, respectively. In order to minimize this effect, we isolated the whole set-up and the temperature was kept stable in a narrow range around 37°C. Despite this, the volume overestimation was consistently present even at small volumes (356 mL) and high flows (4 liters per minute), when heat loss is expected to be minimal.
In our study, we used a TEE matrix probe, as this is a more realistic set-up to perform volume measurements by CEUS during open heart surgery and the perioperative phase. In general the proposed methods are also feasible by TTE, as shown in studies by Mischi [7, 8]. In comparison to TTE, TEE is closer to the heart with minimal ultrasound attenuation in between. Whether the results with a TEE or TTE probe are interchangeable needs to be investigated in future research.
As no calibration is available for the adopted TEE probe in literature, we calibrated the probe by determining the relationship between the SonoVue® concentration and measured acoustic intensity. The calibration was performed at two different temperatures, namely ambient room temperature (20°C) and the typical temperature used for cold thermodilution (4°C). To this purpose, we diluted SonoVue® in saline at 4°C to obtain sufficient signal-to-noise ratio for the thermodilution IDC and because this temperature is routinely used for clinical thermodilution measurements. We found UCA bubbles to be stable longer at lower temperatures , however at the cost of a reduced echogenicity (Figure 2). This is in line with the reported decrease in bubble stability at higher temperatures [17, 18]. At higher temperature the UCA bubbles expand 5% by gas expansion and show increased acoustic backscatter . Our calibration showed no attenuation below 1.0 mg/L at 4°C. Higher doses will produce a non-stationary (concentration dependent) shadow effect that will influence the IDC quantification, possibly affecting the MTT. We calculated our diluting volume for SonoVue® as the total volume in the circuit, which was over 1 liter. Nevertheless at the lowest volume of the tube network (356 mL) and highest flow (4 L/min) we registered some attenuation in the IDC, probably due to a low effective diluting volume of less than 1 L. In spite of this, the MTT reproducibility at these settings was high with an intraclass-correlation of ICC = 0.99 and an average difference with the true volumes of +2 mL for CEUS and for thermodilution +74 mL (Figure 4). Therefore, the influence of attenuation on the MTT assessment seems negligible.
The IDCs are fitted using dedicated models that can possibly influence volume measurement thus leading to over- and/or underestimation. With this respect, the LDRW model seems superior as the fitting is based on both the ascending and descending slopes of the IDC which makes it less sensitive to noise. Still, this may lead to underestimation of the IDC tail compared to more common models like mono-exponential and power-law model . In a study of Ugander et al. magnetic resonance imaging was evaluated as a method for estimating pulmonary blood volume . An in-vitro validation of pulmonary blood volume measurements was carried out, showing a mean difference between the measured volumes and true volumes of 10 ± 2% for the peak-to-peak method and 4 ± 3% for the center of gravity method . The center of gravity and peak-to-peak methods do not use model fitting for the MTT estimation, and they are more sensitive to low signal-to-noise ratios and contrast recirculation. Moreover, they do not provide a physical interpretation of the investigated convective diffusion process, as provided by the LDRW model . The LDRW model provides better fits of skewed IDCs , which are present at high flows and small volumes. Our mean difference for CEUS and the true volumes was −6.5 ± 2.8% using the LDRW double fit method and −8.5 ± 2.9% using LDRW impulse response method. A volume underestimation of −3.3 ± 2.3% was also found by different models and settings in another in-vitro study for volume quantification with magnetic resonance imaging . In particular a slightly lower accuracy and volume underestimation by the impulse response method was reported; however, this was not confirmed by intra-thoracic blood volume measurements in the volunteers. The advantage of the impulse response method consists of making the measurement robust to recirculation and variations in the injection function. Moreover, the identification of the full transpulmonary dilution impulse response brings additional information, possibly adding diagnostic value to the analysis .
Another finding to be discussed relates to the volume underestimation by CEUS. This is likely to be explained with the bubble transport kinetics. It has been reported that bubbles, especially for laminar flow (Reynolds < 2000), show a velocity profile that differs from that of the carrier fluid, leading to a shorter MTT. In particular, while the carrier fluid shows a typical parabolic flow profile, bubbles are reported to travel with a “flatter” profile, whose average velocity over the tube cross section is higher than that of the carrier fluid [23, 24]. Additional explanations, to be verified in future studies, might relate to the concentration profile of bubbles across the tube.
In-vivo studies have another carrier fluid, blood, which could influence the indicator dilution curve. As blood is a more viscous carrier fluid than water, the Reynolds number will decrease. As a result, the diffusion coefficient is expected to be lower. The diffusion coefficient influences the parameter λ of the LDRW model, which equals the Peclet number divided by 2. The Peclet number represents the ratio between diffusion and convection time. An increase in viscosity produces therefore an increase in λ, whereas μ, which is representative of the MTT, is not affected .
Our study may provide a good alternative for volume measurement in the perioperative setting and in critical care. The safety of SonoVue® has been investigated in different settings. In a large retrospective study on assessment of adverse events in 28 Italian Centers, serious adverse events were found in 0.0086% [26, 27]. SonoVue® microbubbles are composed of SF6 gas with a phospholipid monolayer shell. The elimination of the SF6 gas via the lung is reported to be, even in patients with obstructive pulmonary disease and pulmonary fibrosis, in the same range as in healthy volunteers with a 80-90% clearance within 11 minutes . The phospholipid monolayer is metabolized in the liver. These monolayers are commonly used in the formulation and manufacturing of liposomes, a drug delivery system that is approved by the United States Food and Drug Administration (FDA) and considered biologically safe . Even in blunt abdominal trauma patients, the use of SonoVue® was proven to be safe . Only in patients with a recent cardiac infarction or heart failure class III and IV, the European Medicines Agency (EMA) took precautions and these conditions are contraindicated for use of SonoVue®.
Limitations of our study relate to the thermodilution technique. Even with extended isolation of the set-up, volumes were overestimated using thermodilution. The pressure wires are normally used in coronary arteries to measure flow with thermodilution. These wires are very thin and ideal in our in-vitro set-up but not used for larger blood volume measurement. Thus, this correlation cannot be extrapolated in-vivo. Further research will be needed to compare CEUS and evaluate the margin of error with clinically used thermodilution methods for volume measurement, such as PiCCO® (Pulsion Medical Systems, Munich, Germany), and to estimate its value as a minimally-invasive and bedside-applicable technique in the ICU and operating room [3, 30].
CEUS seems a promising, minimally-invasive technique to measure volumes. Our in-vitro measurements showed a good correlation and level of agreement between CEUS volumes and the true volumes. We found a general overestimation of the measured volumes by the thermodilution technique and a general underestimation by CEUS, the latter being more evident for larger volumes and lower flows. This study suggests the use of CEUS to be superior to thermodilution and a good equivalent to the gold standard. We believe that this novel, minimally-invasive and non-nuclear method for measuring blood volume can be an asset in clinical research and practice. However, it is mandatory to validate this novel technique with frequently used techniques in daily, clinical practice.
- Pinsky MR, Payen D: Functional hemodynamic monitoring. Crit Care. 2005, 9: 566-572.View ArticlePubMedPubMed CentralGoogle Scholar
- Swan HJ, Ganz W, Forrester J, Marcus H, Diamond G, Chonette D: Catheterization of the heart in man with use of a flow-directed balloon-tipped catheter. N Engl J Med. 1970, 283: 447-451.View ArticlePubMedGoogle Scholar
- Sakka SG, Ruhl CC, Pfeiffer UJ, Beale R, McLuckie A, Reinhart K, Meier-Hellmann A: Assessment of cardiac preload and extravascular lung water by single transpulmonary thermodilution. Intensive Care Med. 2000, 26: 180-187.View ArticlePubMedGoogle Scholar
- Reuter DA, Huang C, Edrich T, Shernan SK, Eltzschig HK: Cardiac output monitoring using indicator-dilution techniques: basics, limits, and perspectives. Anesth Analg. 2010, 110: 799-811.View ArticlePubMedGoogle Scholar
- Brands J, Vink H, Van Teeffelen JW: Comparison of four mathematical models to analyze indicator-dilution curves in the coronary circulation. Med Biol Eng Comput. 2011, 49: 1471-1479.View ArticlePubMedPubMed CentralGoogle Scholar
- Maddison B, Wolff C, Findlay G, Radermacher P, Hinds C, Pearse RM: Comparison of three methods of extravascular lung water volume measurement in patients after cardiac surgery. Crit Care. 2009, 13: R107-View ArticlePubMedPubMed CentralGoogle Scholar
- Mischi M, Kalker TA, Korsten EH: Contrast echocardiography for pulmonary blood volume quantification. IEEE Trans Ultrason Ferroelectr Freq Control. 2004, 51: 1137-1147.View ArticlePubMedGoogle Scholar
- Mischi M, Jansen AH, Korsten HH: Identification of cardiovascular dilution systems by contrast ultrasound. Ultrasound Med Biol. 2007, 33: 439-451.View ArticlePubMedGoogle Scholar
- Korsten HH, Mischi M, Grouls RJ, Jansen A, van Dantzig JM, Peels K: Quantification in echocardiography. Semin Cardiothorac Vasc Anesth. 2006, 10: 57-62.View ArticlePubMedGoogle Scholar
- Schreiner BF, Murphy GW, James DH, Yu PN: Pulmonary blood volume in patients with congestive heart failure. Trans Assoc Am Physicians. 1966, 79: 250-261.PubMedGoogle Scholar
- McGaff CJ, Roveti GC, Glassman E, Milnor WR: Pulmonary blood volume in rheumatic heart disease and its alteration by isoproterenol. Circulation. 1963, 27: 77-84.View ArticlePubMedGoogle Scholar
- Wise ME: Tracer dilution curves in cardiology and random walk and lognormal distributions. Acta Physiol Pharmacol Neerl. 1966, 14: 175-204.PubMedGoogle Scholar
- Bland JM, Altman DG: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet. 1986, 1: 307-310.View ArticlePubMedGoogle Scholar
- Shrout PE, Fleiss JL: Intraclass correlations: uses in assessing rater reliability. Psychol Bull. 1979, 86: 420-428.View ArticlePubMedGoogle Scholar
- Weir JP: Quantifying test-retest reliability using the intraclass correlation coefficient and the SEM. J Strength Cond Res. 2005, 19: 231-240.PubMedGoogle Scholar
- Herold I, Kuenen M, Mischi M, Korsten H: Blood volume and ejection fraction measurements using CEUS. Proceedings on the 15th European Symposium on Ultrasound Contrast Imaging. 2010, 55-56. Erasmus University RotterdamGoogle Scholar
- Mulvana H, Stride E, Hajnal JV, Eckersley RJ: Temperature dependent behavior of ultrasound contrast agents. Ultrasound Med Biol. 2010, 36: 925-934.View ArticlePubMedGoogle Scholar
- Goddi A, Novario R, Tanzi F, Di Liberto R, Nucci P: In vitro analysis of ultrasound second generation contrast agent diluted in saline solution. Radiol Med. 2004, 107: 569-579.PubMedGoogle Scholar
- Vos HJ: Single microbubble imaging Rotterdam. 2010, Rotterdam: Erasmus University, 131-145.Google Scholar
- Ugander M, Kanski M, Engblom H, Gotberg M, Olivecrona GK, Erlinge D, Heiberg E, Arheden H: Pulmonary blood volume variation decreases after myocardial infarction in pigs: a quantitative and noninvasive MR imaging measure of heart failure. Radiology. 2010, 256: 415-423.View ArticlePubMedGoogle Scholar
- Mischi M, den Boer JA, Korsten HH: On the physical and stochastic representation of an indicator dilution curve as a gamma variate. Physiol Meas. 2008, 29: 281-294.View ArticlePubMedGoogle Scholar
- Mischi M, van den Bosch HC, den Boer JA, Verwoerd J, Grouls RJ, Peels CH, Korsten HH: Intra-thoracic blood volume measurement by contrast magnetic resonance imaging. Magn Reson Med. 2009, 61: 344-353.View ArticlePubMedGoogle Scholar
- Keller MW, Segal SS, Kaul S, Duling B: The behavior of sonicated albumin microbubbles within the microcirculation: a basis for their use during myocardial contrast echocardiography. Circ Res. 1989, 65: 458-467.View ArticlePubMedGoogle Scholar
- Tangelder GJ, Slaaf DW, Muijtjens AM, Arts T, oude Egbrink MG, Reneman RS: Velocity profiles of blood platelets and red blood cells flowing in arterioles of the rabbit mesentery. Circ Res. 1986, 59: 505-514.View ArticlePubMedGoogle Scholar
- Mischi M, Kalker AA, Korsten HH: Cardiac image segmentation for contrast agent videodensitometry. IEEE Trans Biomed Eng. 2005, 52: 277-286.View ArticlePubMedGoogle Scholar
- Morel DR, Schwieger I, Hohn L, Terrettaz J, Llull JB, Cornioley YA, Schneider M: Human pharmacokinetics and safety evaluation of SonoVue, a new contrast agent for ultrasound imaging. Invest Radiol. 2000, 35: 80-85.View ArticlePubMedGoogle Scholar
- Piscaglia F, Bolondi L: The safety of Sonovue in abdominal applications: retrospective analysis of 23188 investigations. Ultrasound Med Biol. 2006, 32: 1369-1375.View ArticlePubMedGoogle Scholar
- Kang ST, Yeh CK: Ultrasound microbubble contrast agents for diagnostic and therapeutic applications: current status and future design. Chang Gung Med J. 2012, 35: 125-139.PubMedGoogle Scholar
- Cagini L, Gravante S, Malaspina CM, Cesarano E, Giganti M, Rebonato A, Fonio P, Scialpi M: Contrast enhanced ultrasound (CEUS) in blunt abdominal trauma. Crit Ultrasound J. 2013, 5 (1): S9-View ArticlePubMedPubMed CentralGoogle Scholar
- Linton RA, Band DM, Haire KM: A new method of measuring cardiac output in man using lithium dilution. Br J Anaesth. 1993, 71: 262-266.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.