- Open Access
- Open Peer Review
Feasibility and reference values of left atrial longitudinal strain imaging by two-dimensional speckle tracking
Cardiovascular Ultrasound volume 7, Article number: 6 (2009)
The role of speckle tracking in the assessment of left atrial (LA) deformation dynamics is not established. We sought to determine the feasibility and reference ranges of LA longitudinal strain indices measured by speckle tracking in a population of normal subjects.
In 60 healthy individuals, peak atrial longitudinal strain (PALS) and time to peak longitudinal strain (TPLS) were measured using a 12-segment model for the left atrium. Values were obtained by averaging all segments (global PALS and TPLS) and by separately averaging segments measured in the two apical views (4- and 2-chamber average PALS and TPLS).
Adequate tracking quality was achieved in 97% of segments analyzed. Inter and intra-observer variability coefficients of measurements ranged between 2.9% and 5.4%. Global PALS was 42.2 ± 6.1% (5–95° percentile range 32.2–53.2%), and global TPLS was 368 ± 30 ms (5–95° percentile range 323–430 ms). The 2-chamber average PALS was slightly higher than the 4-chamber average PALS (44.3 ± 6.0% vs 40.1 ± 7.9%, p < 0.0001), whereas no differences in TPLS were found (p = 0.93).
Speckle tracking is a feasible technique for the assessment of longitudinal myocardial LA deformation. Reference ranges of strain indices were reported.
The left atrium plays as a booster pomp during late ventricular diastole, as a reservoir for the inflow volume received from pulmonary veins during ventricular systole and isovolumic relaxation, and as a passive conduit during early ventricular diastole and diastasis [1, 2]. Although estimates of left atrial (LA) function can be obtained by two-dimensional echocardiography, Doppler analysis of transmitral and pulmonary vein flow, and Tissue Doppler (TD) assessment of LA myocardial velocities [3–5], quantification of effective LA function still remains a challenging issue. Assessment of atrial deformation profiles obtained using Doppler-derived strain imaging has been recently proposed as an alternative method of exploring LA function. However, this approach is limited by a number of potential drawbacks, including suboptimal reproducibility, angle dependence, and the confounding effect of noise artifacts.
Two-dimensional strain imaging is an echocardiographic technique that uses standard B-mode images for speckle tracking analysis. The speckle pattern (acoustic backscatter generated by the reflected ultrasound beam) is followed frame-by-frame, using a statistical approach based on the detection of the best matching area. The displacement of this speckled pattern is considered to follow myocardial movement, and a change between speckles is assumed to represent myocardial deformation[8, 9].
The aims of this study were to define the feasibility of speckle tracking-based strain imaging for the evaluation of LA wall deformation in a population of healthy subjects, and to identify normality ranges for corresponding strain values.
Sixty consecutive adult healthy subjects, referring to our Echo Laboratory for a diagnostic examination, were included in the study group. All had unremarkable clinical history and normal findings at physical examination, ECG, and baseline echocardiography, and none of them was taking cardiac medications. All subjects gave their written informed consent for participation in the study.
Echocardiographic studies were performed using a high-quality echocardiograph (Vivid 7, GE, USA). Subjects were studied in the left lateral recumbent position. Measurements of left ventricular (LV) and LA dimensions were made in accordance with current American Society of Echocardiography recommendations. LV ejection fraction was measured using the modified biplane Simpson's rule. The ratio between peak early (E) and late (A) diastolic LV filling velocities and E wave deceleration time were determined by standard Doppler imaging. The timings of mitral and aortic valve opening and closure were defined by pulsed wave Doppler tracings of mitral inflow and LV outflow.
For speckle tracking analysis, apical four- and two-chamber views images were obtained using conventional two-dimensional gray scale echocardiography, during breath hold with a stable ECG recording. Particular attention was given to obtain an adequate gray scale image, allowing reliable delineation of myocardial tissue and extracardiac structures. Three consecutive heart cycles were recorded and averaged. The frame rate was set between 60 and 80 frames per second. These settings are recommended to combine temporal resolution with adequate spatial definition, and to enhance the feasibility of the frame-to-frame tracking technique.
Recordings were processed using an acoustic-tracking software (Echo Pac, GE, USA), allowing off-line semi-automated analysis of speckle-based strain[14, 15] (Figure 1). Briefly, LA endocardial surface is manually traced in both four- and two-chamber views by a point-and-click approach. An epicardial surface tracing is then automatically generated by the system, thus creating a region of interest (ROI). After manual adjustment of ROI width and shape, the software divides the ROI into 6 segments, and the resulting tracking quality for each segment is automatically scored as either acceptable or non-acceptable, with the possibility of further manual correction. Segments in which no adequate image quality can be obtained are rejected by the software and excluded from the analysis. Lastly, the software generates strain curves for each atrial segment. In subjects with adequate image quality, a total of 12 segments were then analyzed. To trace the ROI in the discontinuity of LA wall corresponding to pulmonary veins and LA appendage, the direction of LA endocardial and epicardial surfaces at the junction with these structures was extrapolated. Peak atrial longitudinal strain (PALS) was calculated by averaging values observed in all LA segments (global PALS), and by separately averaging values observed in 4- and 2-chamber views (4- and 2-chamber average PALS) (Figure 2). The time to peak longitudinal strain (TPLS) was also measured as the average of all 12 segments (global TPLS) and by separately averaging values observed in the two apical views (4- and 2-chamber average TPLS). In patients in whom some segments were excluded because of the impossibility of achieving adequate tracking, PALS and TPLS were calculated by averaging values measured in the remaining segments.
Reproducibility of PALS and TPLS measurements was assessed in 20 randomly selected subjects. Intra and inter-observer variability coefficients were calculated using images independently recorded in two different occasions by the same investigator or by two different observers.
Data are shown as mean ± SD. Inter- and intra-observer reproducibility was assessed by calculating variability coefficients. Reference values were expressed as mean ± SD and 5–95° percentile ranges. Comparisons were performed using the Student's t test for paired data. A P value < 0.05 was considered statistically significant. Analyses were performed using the SPSS (Statistical Package for the Social Sciences, Chicago, Illinois) software Release 11.5.
Table 1 shows the clinical and echocardiographic characteristics of the study population. Among a total of 720 segments, the software was able to track 697 (96.9%) segments. Adequate tracking of all 12 LV segments was achieved in 50 (83.3%) subjects, and in no cases the number of segments adequately explored was < 8. Average post-processing time per patient was 2 ± 1 min. Inter-observer variability coefficients of global PALS and TPLS were 3.4%, and 4.8%, respectively. For intra-observer variability, the corresponding variability coefficients were 2.9% and 3.8%.
When the reproducibility was separately considered in the two apical views, inter-observer variability coefficients were 4.3% and 4.6% for 4- and 2-chamber average PALS, and 5.4% and 5.3% for TPLS, respectively. Corresponding intra-observer variability coefficients were 3.6%, 4.0%, 4.5%, and 4.8%.
Mean ± SD and 5–95° percentile ranges of global, 4-chamber, and 2-chamber PALS and TPLS observed in the study population are reported in Table 2. Notably, 2-chamber average PALS was significantly higher than 4-chamber average PALS (p < 0.0001), whereas there was no difference between 2- and 4-chamber average TPLS (p = 0.93).
In this study, speckle tracking imaging was found to be a feasible and reproducible method to assess LA longitudinal strain in healthy subjects. The reproducibility of measurements was good, with lower variability in comparison with that obtained by Doppler-derived LA strain imaging. These data suggest that speckle tracking may be considered a promising tool to explore LA myocardial deformation dynamics. Data observed in our population were also used to provide reference ranges for speckle-based LA strain indices.
The strain curve evaluated by either Doppler method or speckle tracking imaging closely follows LA physiology (Figure 3, Additional file 1). During the period of LA reservoir (corresponding to the phases of LV isovolumic contraction, ejection, and isovolumic relaxation), LA strain increases, achieving a peak at the end of LA filling from the venous district, just before mitral valve opening. During the conduit phase, LA strain decreases, showing a plateau during diastasis, and achieving a negative peak at the end of LA contraction. Considering the limitations of classical indices of LA function, assessment of LA strain by speckle tracking may represent a relatively rapid and easy-to-perform technique to explore LA function. This approach may be of potential clinical importance in a number of pathophysiologic conditions typically associated to abnormal LA function, e.g. mitral valve diseases, supraventricular arrhythmias, systemic hypertension, ischemic heart disease, heart failure, atrial stunning, and cardiomyopathies.
In our population, average peak longitudinal strain was higher in apical 2-chamber view than in apical 4-chamber view. It can be hypothesized that the confounding effect of atrial septum and pulmonary veins ostia in the 4-chamber view – two zones where strain profile is abnormal – may have played a role in determining this discrepancy. In contrast, the average time to peak longitudinal strain was similar in the two apical views. Interestingly, these findings are in contrast with a recent report on LA strain measured by the Doppler-derived technique, which found similar peak strain values among different LA walls, but different timing measures. These discrepancies should be evaluated in further studies. It should be noted that in contrast to Doppler-derived strain imaging, speckle tracking has the advantage of being angle-independent, and to be less affected by reverberations, side lobes and drop out artifacts. Nonetheless, intrinsic limitations of speckle tracking include strict frame rate dependency, potential errors in epicardial/endocardial border tracing in subjects with suboptimal image quality, and need for an appropriate learning curve to achieve adequate experience in using analysis softwares.
Some limitations should be considered. The potential difficulty of accurately obtaining a region of interest close enough to the effective shape of the left atrium, and the risk of contamination by signal components arising from structures surrounding the left atrium should also be considered. Although the post-processing time in this study was relatively short, it closely depends on the sonographer's experience. Lastly, because a dedicated software for LA strain analysis has not yet been released, we used the current software for LV analysis to study the LA pattern strain. Future evolutions in this regard may be useful to improve tracking quality of LA myocardial deformation, and to provide a better instrument for the study of LA function. Lastly, it should be emphasized that as for other echocardiographic new technologies, speckle tracking is progressively entering the clinical practice despite no definite data regarding reference ranges and no clear demonstration of clear additive value in particular clinical conditions. The results of this study may contribute to partially fill this gap giving insight on the potential application of speckle tracking to the study of LA function, but further larger analyses are needed.
In summary, speckle tracking can be considered a feasible and reproducible technique for the assessment of LA longitudinal deformation dynamics. Normal values of longitudinal strain indices were reported.
Pagel PS, Kehl F, Gare M, Hettrick DA, Kersten JR, Warltier DC: Mechanical function of the left atrium. New insights based on analysis of pressure-volume relations and Doppler echocardiography. Anestesiology. 2003, 98: 975-994. 10.1097/00000542-200304000-00027.
Di Salvo G, Galderisi M, Rea A, Ansalone G, Dini FL, Gallina S, Mele D, Montisci R, Sciomer S, Mondillo S, Di Bello V, Marino PN, Gruppo di Lavoro di Ecocardiografia; Società Italiana di Cardiologia: Evaluation of atrial function by echocardiography. G Ital Cardiol. 2007, 8: 225-235.
Nakao F, Wasaki Y, Kimura M, Iwami T, Iida H, Wakeyama T, Miura T, Ogawa H, Matsuzaki M: Evaluation of left atrial function by the functional volume change curve derived from Doppler flow spectra. Jpn Circ J. 2001, 65: 953-7. 10.1253/jcj.65.953.
Marino P, Faggian G, Bertolini P, Mazzucco A, Little W: Early mitral deceleration and left atrial stiffness. Am J Physiol Heart Circ Physiol. 2004, 287: H1172-1178. 10.1152/ajpheart.00051.2004.
Pérez-Paredes M, Gonzálvez M, Ruiz Ros JA, Giménez DM, Carnero A, Carrillo A, Cubero T, Martínez-Corbalán FR, García Almagro F: Assessment of Left Atrial Wall Velocities by Pulsed Wave Tissue Doppler Imaging. A New Approach to the Study of Atrial Function. Rev Esp Cardiol. 2004, 57: 1059-1066. 10.1157/13068167.
Di Salvo G, Caso P, Lo Piccolo R, Fusco A, Martiniello AR, Russo MG, D'Onofrio A, Severino S, Calabró P, Pacileo G, Mininni N, Calabró R: Atrial myocardial deformation properties predict maintenance of sinus rhythm after external cardioversion of recent-onset lone atrial fibrillation: a color Doppler myocardial imaging and transthoracic and transesophageal echocardiographic study. Circulation. 2005, 112: 387-395. 10.1161/CIRCULATIONAHA.104.463125.
D'hooge J, Heimdal A, Jamal F, Kukulski T, Bijnens B, Rademakers F, Hatle L, Suetens P, Sutherland GR: Regional strain and SR measurements by cardiac ultrasound: principles, implementation and limitations. Eur J Echocardiogr. 2000, 1: 154-170. 10.1053/euje.2000.0031.
Teske AJ, De Boeck BW, Melman PG, Sieswerda GT, Doevendans PA, Cramer MJ: Echocardiographic quantification of myocardial function using tissue deformation imaging, a guide to image acquisition and analysis using tissue Doppler and speckle tracking. Cardiovascular Ultrasound. 2007, 5: 27-10.1186/1476-7120-5-27.
Notomi Y, Lysyansky P, Setser RM, Shiota T, Popović ZB, Martin-Miklovic MG, Weaver JA, Oryszak SJ, Greenberg NL, White RD, Thomas JD: Measurement of Ventricular Torsion by Two-Dimensional Ultrasound Speckle Tracking Imaging. J Am Coll Cardiol. 2005, 45: 2034-2041. 10.1016/j.jacc.2005.02.082.
Di Salvo G, Drago M, Pacileo G, Rea A, Carrozza M, Santoro G, Bigazzi MC, Caso P, Russo MG, Carminati M, Calabro' R: Atrial function after surgical and percutaneous closure of atrial septal defect: a strain rate imaging study. J Am Soc Echocardiogr. 2005, 18: 930-933. 10.1016/j.echo.2005.01.029.
D'Andrea A, Caso P, Romano S, Scarafile R, Cuomo S, Salerno G, Riegler L, Limongelli G, Di Salvo G, Romano M, Liccardo B, Iengo R, Ascione L, Del Viscovo L, Calabrò P, Calabrò R: Association between left atrial myocardial function and exercise capacity in patients with either idiopathic or ischemic dilated cardiomyopathy: A two-dimensional speckle strain study. Int J Cardiol.
Lang RM, Bierig M, Devereux RB, Flachskampf FA, Foster E, Pellikka PA, Picard MH, Roman MJ, Seward J, Shanewise JS, Solomon SD, Spencer KT, Sutton MS, Stewart WJ, Chamber Quantification Writing Group; American Society of Echocardiography's Guidelines and Standards Committee; European Association of Echocardiography: Recommendations for Chamber Quantification: A Report from the American Society of Echocardiography's Guidelines and Standards Committee and the Chamber Quantification Writing Group, Developed in Conjunction with the European Association of Echocardiography, a Branch of the European Society of Cardiology. J Am Soc Echocardiogr. 2005, 18: 1440-1463. 10.1016/j.echo.2005.10.005.
Serri K, Reant P, Lafitte M, Berhouet M, Le Bouffos V, Roudaut R, Lafitte S: Global and regional myocardial function quantification by two dimensional strain. J Am Coll Cardiol. 2006, 47: 1175-1181. 10.1016/j.jacc.2005.10.061.
Langeland S, D'hooge J, Wouters PF, Leather HA, Claus P, Bijnens B, Sutherland GR: Experimental validation of a new ultrasound method for the simultaneous assessment of radial and longitudinal myocardial deformation independent of insonation angle. Circulation. 2005, 112: 2157-2162. 10.1161/CIRCULATIONAHA.105.554006.
Kaluzynski K, Chen X, Emelianov SY, Skovoroda AR, O'Donnell M: Strain rate imaging using two-dimensional speckle tracking. IEEE Trans Ultrason Ferroelectr Freq Control. 2001, 48: 1111-1123. 10.1109/58.935730.
Sirbu C, Herbots L, D'hooge J, Claus P, Marciniak A, Langeland T, Bijnens B, Rademakers FE, Sutherland GR: Feasibility of strain and strain rate imaging for the assessment of regional left atrial deformation: a study in normal subjects. Eur J Echocardiogr. 2006, 7: 199-208. 10.1016/j.euje.2005.06.001.
Leung DY, Boyd A, Ng AA, Chi C, Thomas L: Echocardiographic evaluation of left atrial size and function: current understanding, pathophysiologic correlates, and prognostic implications. Am Heart J. 2008, 156: 1056-64. 10.1016/j.ahj.2008.07.021.
The authors wish to thank Vittorio Rastelli and Fabio Cattaneo, GE Healthcare, Milano, Italy, for their important technical support.
The authors declare that they have no competing interests.
MC, MC, ML, and EP were responsible for the collection of data and drafted the manuscript. PB performed the statistical analysis and revised the manuscript for important intellectual content. SM was responsible for the design of the study and revised the manuscript for important intellectual content. EM provided precious support for the acquisition of data in applying the speckle tracking software to the study of left atrial myocardium. MG also revised the manuscript for important intellectual content. All authors read and approved the final manuscript.
Matteo Cameli, Maria Caputo contributed equally to this work.
About this article
Cite this article
Cameli, M., Caputo, M., Mondillo, S. et al. Feasibility and reference values of left atrial longitudinal strain imaging by two-dimensional speckle tracking. Cardiovasc Ultrasound 7, 6 (2009) doi:10.1186/1476-7120-7-6
- Left Atrial
- Longitudinal Strain
- Speckle Tracking
- Mitral Valve Disease
- Left Atrial Function